<?xml version="1.0"?>
<ArrayOfFBEvent xmlns:xsd="http://www.w3.org/2001/XMLSchema" xmlns:xsi="http://www.w3.org/2001/XMLSchema-instance">
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment />
    <StatusString>Edited</StatusString>
    <TimeStamp>2024-08-24T00:12:47Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Jaime Lopez</User>
    <Comment>Verified with DataPRO 4.0.1152, 4.2.411, 4.3.155
 
Tested:
 
With a variety of SLICE family devices (SPS, Base Slice +, S6A, S6A-BR, S6 and S6DB)
 
Created a test setup with these units and verified all recording modes
 
 
 
that allowed for more than 1 event was able to limit the user to only 100 events. By setting to 200, DataPRO would setback to 100 events MAX.
 
Verified also that if user selects TSR AIR as hardware, user is able to program for more than 100 events. ✔</Comment>
    <StatusString>Closed</StatusString>
    <TimeStamp>2024-05-13T19:02:49Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>Closed</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment />
    <StatusString>ReleaseNoted</StatusString>
    <TimeStamp>2024-04-09T17:38:11Z</TimeStamp>
    <ReleaseNote>All DAS are limited to a max of 100 events again</ReleaseNote>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment />
    <StatusString>Edited</StatusString>
    <TimeStamp>2024-04-09T17:37:57Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment />
    <StatusString>Edited</StatusString>
    <TimeStamp>2024-04-09T16:29:41Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment />
    <StatusString>Edited</StatusString>
    <TimeStamp>2024-04-09T16:29:36Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Jaime Lopez</User>
    <Comment />
    <StatusString>Assigned</StatusString>
    <TimeStamp>2024-04-01T22:27:56Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment>4.0/4.2/4.3@50554</Comment>
    <StatusString>Resolved</StatusString>
    <TimeStamp>2024-03-29T22:13:22Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>Resolved</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment />
    <StatusString>Edited</StatusString>
    <TimeStamp>2024-03-29T22:12:53Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment />
    <StatusString>Assigned</StatusString>
    <TimeStamp>2024-03-29T22:04:42Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment />
    <StatusString>Edited</StatusString>
    <TimeStamp>2024-03-29T22:04:33Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Chris Balogh</User>
    <Comment>https://dtsweb.zendesk.com/agent/tickets/13296</Comment>
    <StatusString>Edited</StatusString>
    <TimeStamp>2024-03-29T21:40:29Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Chris Balogh</User>
    <Comment>Workaround is to have customer arm with &lt;=100 events. No way to arm with more than that, so workaround is explanation.

It does open us to lost events, but that's not in the descriptions of the Severity listing.</Comment>
    <StatusString>Edited</StatusString>
    <TimeStamp>2024-03-29T21:39:16Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Jaime Lopez</User>
    <Comment />
    <StatusString>Edited</StatusString>
    <TimeStamp>2024-03-29T21:35:57Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Jaime Lopez</User>
    <Comment />
    <StatusString>Edited</StatusString>
    <TimeStamp>2024-03-29T21:35:39Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Jaime Lopez</User>
    <Comment />
    <StatusString>Edited</StatusString>
    <TimeStamp>2024-03-29T21:34:41Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Jaime Lopez</User>
    <Comment />
    <StatusString>Edited</StatusString>
    <TimeStamp>2024-03-29T21:33:20Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Jaime Lopez</User>
    <Comment />
    <StatusString>Assigned</StatusString>
    <TimeStamp>2024-03-29T21:32:53Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Jaime Lopez</User>
    <Comment>Description:
 
Chris notified me that S6A should only be limited to 100 max events. Set number of events to 255 and armed with no issue. Yet in FWTU the max events I can set to is 100. Upon arming in DataPRO I see the following errors in the logs. 
 
Logs:
 
2024-03-29 14:23:52.804 -&gt;  Attribute - SetSystemAttribute [00000382], S6A0227, local: 192.168.1.4:56629, Remote: 192.168.4.27:8301 Key: MaxEvent, DataType: UInt8, Length: 1, Overwrite: True, Value: 255
 
 2024-03-29 14:23:52.808 &lt;-  Attribute - SetSystemAttribute [00000382]\[00000382] - (3.9889ms), S6A0227 StatusAttributeSetError
 
2024-03-29 14:23:52.808 Exception setting max number of events !!  An exception of type: System.Exception occurred Module: ICommand.dll, Name: InternalSyncExecuteMessage: S6A0227 was executing  DTS.DASLib.Command.SLICE.SetSystemAttribute   and reported SyncExecute: response.Status = StatusAttributeSetError for sequence 382\382.  Please report this information to Technical Support StackTrace:    at DTS.DASLib.Command.AbstractCommandBase.InternalSyncExecute() StackTrace:    at DTS.DASLib.Command.AbstractCommandBase.SyncExecute()    at DTS.DASLib.Service.Slice`1.SetMaxEvents(Int32 numEvents)
 
 
 
This could be related to case 26817</Comment>
    <StatusString>Opened</StatusString>
    <TimeStamp>2024-03-29T21:32:53Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>Opened</Status>
  </FBEvent>
</ArrayOfFBEvent>