<?xml version="1.0"?>
<ArrayOfFBEvent xmlns:xsd="http://www.w3.org/2001/XMLSchema" xmlns:xsi="http://www.w3.org/2001/XMLSchema-instance">
  <FBEvent>
    <User>Jaime Lopez</User>
    <Comment>Verified w/ DataPRO version 4.0.959, 4.1.315
 
Enabled tilt sensor sw temp logging, armed S6DB with external temp sensor connected to AUX connection
 
Armed in DataPRO and took event
 
CSV shows more than one entry and timestamp is correct. (not in UTC though)</Comment>
    <StatusString>Closed</StatusString>
    <TimeStamp>2023-06-08T00:07:19Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>Closed</Status>
  </FBEvent>
  <FBEvent>
    <User>Jaime Lopez</User>
    <Comment />
    <StatusString>Assigned</StatusString>
    <TimeStamp>2023-06-07T16:23:36Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment />
    <StatusString>Edited</StatusString>
    <TimeStamp>2023-05-24T20:57:43Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Jaime Lopez</User>
    <Comment />
    <StatusString>Assigned</StatusString>
    <TimeStamp>2023-05-23T21:45:30Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment />
    <StatusString>ReleaseNoted</StatusString>
    <TimeStamp>2023-04-25T17:50:20Z</TimeStamp>
    <ReleaseNote>now sets RTC clock on S6DB when arming in UTC and expects to receive UTC time from S6DB</ReleaseNote>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment />
    <StatusString>Edited</StatusString>
    <TimeStamp>2023-04-25T17:48:52Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Jaime Lopez</User>
    <Comment />
    <StatusString>Assigned</StatusString>
    <TimeStamp>2023-04-25T15:07:44Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment>4.0/4.1@48585</Comment>
    <StatusString>Resolved</StatusString>
    <TimeStamp>2023-04-21T14:38:32Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>Resolved</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment />
    <StatusString>Edited</StatusString>
    <TimeStamp>2023-04-20T22:48:03Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment />
    <StatusString>Edited</StatusString>
    <TimeStamp>2023-04-20T22:46:23Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment />
    <StatusString>Edited</StatusString>
    <TimeStamp>2023-04-20T21:59:22Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment>I've noticed tilt is setting local while DP is setting UTC currently;
 
 
 
                if (SupportsTimeSynchronization)
                 {
                     SetBaseSystemTime sbst = new SetBaseSystemTime(this);
                     sbst.SystemTime = DateTime.UtcNow; //18361 using UTC time over local time in case of later data processing
                     sbst.SyncExecute();
                 }
 
vs
 
            RealtimeInteruption(() =&gt;
             {
                 var setBaseTime = new SetBaseSystemTime(_comm, 10000) { SystemTime = DateTime.Now };
                 setBaseTime.SyncExecute();
             });
 
I'll need to check that DP is executing this command as well this is what time is being based on.
 
also noticed DP was never setting RTC for S6DB, so now it does.
 
4.0/4.1@48582.
 
Tilt Sensor Control 2.1 now also sets the time in UTC as does DP. it should download in local time in the CSV.
 
for testing, set the S6DB to record temperatures using latest tilt sensor control, then run a test with the S6DB and download all and verify that the temperature download are time stamped in local time. 
 
Additionally you can verify that DataPRO sets the base time in UTC in the logs:
 
2023-04-20 14:40:00.808 -&gt;  Information - SetBaseSystemTime [00000623], S6DB0010, local: 192.168.3.17:50483, Remote: 192.168.4.151:8301 System Time: 4/20/2023 9:39:58 PM
 2023-04-20 14:40:00.812 &lt;-  Information - SetBaseSystemTime [00000623]\[00000623] - (4.0013ms), S6DB0010</Comment>
    <StatusString>Edited</StatusString>
    <TimeStamp>2023-04-20T20:46:08Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment>re-activated, per Nate TiltControl should set it as UTC, dataPRO should expect it in UTC, the local should only be for display/consumption purposes.</Comment>
    <StatusString>Reactivated</StatusString>
    <TimeStamp>2023-04-20T15:42:10Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>Reactivated</Status>
  </FBEvent>
  <FBEvent>
    <User>Jaime Lopez</User>
    <Comment />
    <StatusString>Assigned</StatusString>
    <TimeStamp>2023-04-19T23:27:15Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Jaime Lopez</User>
    <Comment />
    <StatusString>Assigned</StatusString>
    <TimeStamp>2023-04-17T15:34:21Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment />
    <StatusString>Edited</StatusString>
    <TimeStamp>2023-04-13T22:53:09Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment />
    <StatusString>Edited</StatusString>
    <TimeStamp>2023-04-13T22:47:25Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment />
    <StatusString>Edited</StatusString>
    <TimeStamp>2023-04-13T22:47:16Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment>4.0/4.1@48529
 
I couldn't duplicate the issue with only one line showing up, however I did notice the timestamps were wrong.  Looking more into it there appears to be a difference in timestamps, 
 
This could be a firmware or tilt control specific issue !!! however, with tilt control 2.1 and datapro 4.0 it seems like the clock is seeded in local and not UTC, so when read back from the S6DB it should be read in as local, not UTC.  This doesn't seem right, it should be stored in UTC, not local, but that appears to be what TiltControl2.1 is doing I think?  I switched the read from datapro to use local time, which is what tilt control is using as well.
 
related case 29554
 
http://manuscript.dts.local/f/cases/29554/YUBA-Download-CSV</Comment>
    <StatusString>Resolved</StatusString>
    <TimeStamp>2023-04-13T22:46:03Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>Resolved</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment />
    <StatusString>Edited</StatusString>
    <TimeStamp>2023-04-11T15:37:17Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment />
    <StatusString>Assigned</StatusString>
    <TimeStamp>2023-03-28T20:09:56Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment>[12:47 PM] Chad Ivan
 

 I am testing the SPDB with a ATD that is logging temperatures.  I see a temperature folder and inside there are two .CSV files, they only contain a single temperature sample instead of the full temperature log stored in the DB...
 
[12:49 PM] Chad Ivan
 

 Date and time are also not matching.  This test was 2023_03_28 14_39
 

 S6DB30014_638156115794729263.CSV
 

 [1:04 PM] Dan McFadden
 

 I can create an issue and look into it, I can't say offhand why that would be</Comment>
    <StatusString>Opened</StatusString>
    <TimeStamp>2023-03-28T20:09:56Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>Opened</Status>
  </FBEvent>
</ArrayOfFBEvent>