<?xml version="1.0"?>
<ArrayOfFBEvent xmlns:xsd="http://www.w3.org/2001/XMLSchema" xmlns:xsi="http://www.w3.org/2001/XMLSchema-instance">
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment />
    <StatusString>Edited</StatusString>
    <TimeStamp>2023-05-24T20:48:25Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Justin Richardson</User>
    <Comment>Tested with dp  4.0.844.48194 and 4.1.194.48187    verified when changing clock sources in run test and saving that check channels  and trigger check both reboot the unit and shows message" clock sources have changed"</Comment>
    <StatusString>Closed</StatusString>
    <TimeStamp>2023-02-06T23:40:47Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>Closed</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment />
    <StatusString>ReleaseNoted</StatusString>
    <TimeStamp>2023-01-31T19:13:17Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Justin Richardson</User>
    <Comment />
    <StatusString>Assigned</StatusString>
    <TimeStamp>2023-01-30T16:48:40Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>John Dowling</User>
    <Comment>Changes:
 
 
Run Clock Source settings enforcement in Check Channels &amp; Check Trigger tabs, not just Run Test
 
 
Testing:
 
 
Create a test with Clock Source settings other than "None/None". Save and Run Test. Clock sources should be set as normal in Hardware navstep (i.e. get a notice that settings are changing and the unit is rebooting), chosen settings should show in Diagnostics. No need to arm/collect data.
 
Back in Edit Test Setup, choose settings other than the ones initially chosen (user complaint involved changing to None/None). Save and go to Check Channels. Clock sources should now also go through the set routine in the Hardware Navstep and the new settings should show in Diagnostics.
 
Repeat, but with Check Trigger in Step 2.
 
 
Committed to v4.0 maint and v4.1 dev in r48152</Comment>
    <StatusString>Resolved</StatusString>
    <TimeStamp>2023-01-28T02:04:09Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>Resolved</Status>
  </FBEvent>
  <FBEvent>
    <User>John Dowling</User>
    <Comment />
    <StatusString>Assigned</StatusString>
    <TimeStamp>2023-01-25T01:47:16Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>John Dowling</User>
    <Comment />
    <StatusString>Edited</StatusString>
    <TimeStamp>2023-01-24T18:17:10Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>John Dowling</User>
    <Comment />
    <StatusString>Assigned</StatusString>
    <TimeStamp>2023-01-24T18:16:55Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment />
    <StatusString>Edited</StatusString>
    <TimeStamp>2023-01-17T21:51:33Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment />
    <StatusString>Assigned</StatusString>
    <TimeStamp>2023-01-03T22:57:43Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment />
    <StatusString>Edited</StatusString>
    <TimeStamp>2023-01-03T19:11:17Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment />
    <StatusString>Edited</StatusString>
    <TimeStamp>2022-12-13T17:24:21Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment>complaint seems to be that the clock sync profile should have been unset and shouldn't show up in diagnostics - should show None or not show at all.</Comment>
    <StatusString>Edited</StatusString>
    <TimeStamp>2022-12-13T16:33:44Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment />
    <StatusString>Edited</StatusString>
    <TimeStamp>2022-12-13T16:33:07Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment />
    <StatusString>Assigned</StatusString>
    <TimeStamp>2022-12-13T16:33:00Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment>Hi Dan
  
 In a previous test I configured my S6A with GPS + 1PPS clock sync..
  
  
 2022-12-13 15:43:02.050 Entering ConfigurationService.SetClocks on thread 7 with units: (S6A0083) Called from: CheckHardware.DoWork Line: 0
 2022-12-13 15:43:02.062 -&gt;  Attribute - SetSystemAttribute [00000072], S6A0083, local: 192.168.0.252:62997, Remote: 192.168.0.83:8301 Key: ClockSyncProfile, DataType: UInt8, Length: 1, Overwrite: True, Value: 24
 2022-12-13 15:43:02.065 &lt;-  Attribute - SetSystemAttribute [00000072]\[00000072] - (3ms), S6A0083
 2022-12-13 15:43:02.065 S6A0083 has completed service ConfigurationService
 2022-12-13 15:43:02.065 ConfigurationService finished - total time (0m 0s 15ms)
 2022-12-13 15:43:02.065 ConfigurationService.SetClocks: All units are now finished, calling ServiceAvailable
 2022-12-13 15:43:02.067 Entering ConfigurationService.Reboot on thread 7 with units: (S6A0083) Called from: CheckHardware.DoWork Line: 0
 2022-12-13 15:43:02.072 -&gt;  FirmwareUpdate - Reboot [00000073], S6A0083, local: 192.168.0.252:62997, Remote: 192.168.0.83:8301
 2022-12-13 15:43:02.073 &lt;-  FirmwareUpdate - Reboot [00000073]\[00000073] - (0.9998ms), S6A0083
 2022-12-13 15:43:02.073 S6A0083 has completed service ConfigurationService
 2022-12-13 15:43:02.074 ConfigurationService finished - total time (0m 0s 6ms)
 2022-12-13 15:43:02.075 ConfigurationService.Reboot: All units are now finished, calling ServiceAvailable
 2022-12-13 15:43:02.075 BeginDisconnect 192.168.0.83:8301
 2022-12-13 15:43:02.077 ModalWindow: The Clock Source and/or UART settings have changed on the following unit(s). Settings will be updated and the unit(s) will be rebooted. Please wait a moment and check hardware again. S6A0083
 2022-12-13 15:43:02.078 EndDisconnect() 192.168.0.83:8301
  
 But now in my next test I have no clock sync.,.
  
 
  
 
  
 But Check Channels doesn't run the any clock sync profile adjust ..
  
 
  
 I have to go to Run Test to clear the error but that is nasty as I just want to check out my sensors ..
  
 
 
 
 
 
 
 
 
 
 
 
 
 
 
DataPRO
 
LOG
 
 
 
 
 
 
 
 
 
 
 
DataPRO 2022-12-13 16.17.23.log
 
BZ2</Comment>
    <StatusString>Opened</StatusString>
    <TimeStamp>2022-12-13T16:32:59Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>Opened</Status>
  </FBEvent>
</ArrayOfFBEvent>