<?xml version="1.0"?>
<ArrayOfFBEvent xmlns:xsd="http://www.w3.org/2001/XMLSchema" xmlns:xsi="http://www.w3.org/2001/XMLSchema-instance">
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment />
    <StatusString>ReleaseNoted</StatusString>
    <TimeStamp>2025-04-01T20:59:40Z</TimeStamp>
    <ReleaseNote>Corrects an issue where ROI downloaded was shorter than expected when using multiple sample rates</ReleaseNote>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Jaime Lopez</User>
    <Comment>Verified fix with DP v4.0.1208, 4.2.547, 4.4.236
 
Created a test with 2 or more DAS. Mixed sampling rate 
 
1 at 10k
 
1 at 100k
 
common status line disabled
 
event length for 90 seconds. 
 
multiple ROIxChannels
 
Ran test and download ROI</Comment>
    <StatusString>Closed</StatusString>
    <TimeStamp>2025-04-01T20:58:51Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>Closed</Status>
  </FBEvent>
  <FBEvent>
    <User>Jerry Lawrence</User>
    <Comment>https://dts.box.com/s/mnxl93lhq34largpc1amdvcvhr3kxulk</Comment>
    <StatusString>Edited</StatusString>
    <TimeStamp>2025-03-31T21:56:39Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Jaime Lopez</User>
    <Comment />
    <StatusString>Assigned</StatusString>
    <TimeStamp>2025-03-31T21:48:28Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Jerry Lawrence</User>
    <Comment />
    <StatusString>Edited</StatusString>
    <TimeStamp>2025-03-31T20:42:13Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Jerry Lawrence</User>
    <Comment>52271 is 4.0; 52272 is 4.2; 52273 is 4.4
 
To test:
 
Import the attached 4.2 Test Setup (it contains multiple DAS, and assigns a lower sample rate to a DAS that's lower in the list of DAS, than a DAS with a higher sample rate).
 
Run the test for 90 seconds, and on the Download ROI step, ensure that the "ROI period end (sec)" value is 65, and not approx. 45.</Comment>
    <StatusString>Resolved</StatusString>
    <TimeStamp>2025-03-31T20:41:33Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>Resolved</Status>
  </FBEvent>
  <FBEvent>
    <User>Jerry Lawrence</User>
    <Comment />
    <StatusString>Assigned</StatusString>
    <TimeStamp>2025-03-31T19:19:04Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment>possibly related case 43682</Comment>
    <StatusString>Edited</StatusString>
    <TimeStamp>2025-03-31T15:48:38Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment />
    <StatusString>Assigned</StatusString>
    <TimeStamp>2025-03-31T15:32:32Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment>VPG Security WARNING: This is an external email. Do not click links or open attachments unless you recognize the sender and know the content is safe.
     

  
 
##- Please type your reply above this line -##
 
You are a follower on this request (#14994, #dtsweb.zendesk.com/agent/tickets/14994). Reply to this email to add an internal note to the request.
         
   
Kate Martin (DTS)
 
Mar 31, 2025, 15:24 GMT+1
 
Interestingly 4.2.526 appears to not have this issue ..or at least I can't replicate my theory in this version.
  
 
  
         
 
         
   
Kate Martin (DTS)
 
Mar 31, 2025, 14:08 GMT+1
 
Hi Dan, Jerry
  
 Thank you for the other "good" and "bad" tests Jordan
  
 I think I have finally twigged what is happening here - not sure why I missed it last week sorry.
  
 Here is my 90secs post T0 hybrid test set up with ROI 2 set to 65 secs...
  
  
  
 
  
  
 The key to replicating Dan is to have the DAS that is being used in ROI_2 sampling at a higher rate than the DAS in ROI_1.
  
 That was what was the delta in the 2 MIRA tests set ups - one had ATD's sampling at 10,000sps and the SPS at 20.000sps and ther "good" one had everything at 20,000sps.
  
 With the ROI_2 DAS sampling at equal to or lower than the main ROI_1 DAS all is well..
  
 
  
 But switch it around like I have here with SPS00254 (the ROI_2 DAS) sampling at 20,000 
  
  
 
  
 and even thought the logs files correctly report "total samples recorded" and "trigger sample number" for example, it is the GUI that for some reason recomputes the ROI for the DAS in ROI_2 for some reason ..
  
 I suspect maybe it is using the "total samples recorded" from the ROI_1 DAS and appling the ROI_2 DAS sample rate to that number to come up with the highly suspect 45 seconds which is 90 secs total test duration divided by "DAS ROI 2 sample rate". 
  
 Does the computation for adjusting the available ROI respect the sampling rates for all DAS in that ROI ?
  
 
  
  
  
  
 
  
 logs and test set up folder attached Dan.
  
 Thxs
  
 
Attachment(s)
 DataPRO 2025-03-31 13.49.48.log.bz2
 DataPRO 2025-03-31 13.50.24.log.bz2
 2025_03_31 13_44.7z
         
 
         
   
Kate Martin (DTS)
 
Mar 31, 2025, 12:29 GMT+1
     
Private note
     
Conclusion
  
 Good Test B7104 has all DAS at 20,000sps
  
 Bad Test B7106 has mixed DAS at 10,000sps and 20,000sps
  
  
         
 
         
   
Kate Martin (DTS)
 
Mar 31, 2025, 12:16 GMT+1
     
Private note
     
.dts file
  
 
  
  
         
 
         
   
Kate Martin (DTS)
 
Mar 31, 2025, 11:56 GMT+1
     
Private note
     
.xml files that is 
         
 
         
   
Kate Martin (DTS)
 
Mar 31, 2025, 11:55 GMT+1
     
Private note
     
Only ROI Period 2 references in .dts files look good..
  
 Good Test B7104
  
 
  
 "Bad Test"
  
 
  
  
         
 
         
   
Jordon Tray
 
Mar 17, 2025, 08:21 GMT
 
Good Morning Kate,
 
 
 
Any advancements on this? Colin keen to work out the issue as we have many EuroNCAP tests in the pipeline.
 
 
 
Thanks,
 
 
     
Jordan Tray
 Engineering Technician
     
         
   
   
   
   
   
             
Direct Tel:  
   
+44 24 7635 5000
     
Email:  
   
jordan.tray@horiba-mira.com
         
HORIBA MIRA Ltd.
 Watling Street, Nuneaton
 Warwickshire, CV10 0TU, UK
     
www.horiba-mira.com
 www.vehicletesting.solutions
 www.miratechnologypark.com
     

 If you are visiting one of our sites please use our visitor information guide for Nuneaton  and Quatro Park.
 
Attachment(s)
 image021.png
 image020.jpg
 image024.png
 image027.png
 image028.png
 image029.png
 image032.png
 image033.png
 image034.png
 image035.png
 image036.png
 image037.png
 image025.jpg
 image026.png
 image030.png
 image031.png
         
 
         
   
Dan McFadden (DTS)
 
Mar 10, 2025, 15:20 GMT
     
Private note
     
Is it possible it somehow picked up a configuration file from an unexpected location?  Maybe it a corrupted one on the unit so it used a backup on the PC and it happened to have a backup on the pc that was old for some reason?
 
 
                         
                         
       
       
                                           
Dan​​​​
   
 
   
Mcfadden
             
Software Engineering Manager
                     
Diversified Technical Systems, Inc.
                     
e:
       
Dan.Mcfadden@dtsweb.com
       
|
           
o:
       
+1 562 493 0158 x137
       
|
       
www.dtsweb.com
                                         
CONFIDENTIALITY: This email and any attachments may contain information from Diversified Technical Systems, Inc. (DTS) which​ is confidential ​and/or proprietary. If you received this transmission in error, any use, reproduction or dissemination of this transmission​ is strictly prohibited. ​Please delete this message and notify sender immediately.  
 ​EXPORT CONTROL NOTICE: This e-mail may contain technical data whose export, transfer, and/or disclosure may be controlled by​ the US ​International Traffic in Arms Regulation (ITAR) 22 CFR part 120-130 or the Export Administration Regulations (Commerce).
 ​
 ​
                   
 
                 
Attachment(s)
 ~WRD0004.jpg
         
 
         
   
Jordon Tray
 
Mar 7, 2025, 13:48 GMT
 
Attached here is working test..
 
 
     
Jordan Tray
 Engineering Technician
     
         
   
   
   
   
   
             
Direct Tel:  
   
+44 24 7635 5000
     
Email:  
   
jordan.tray@horiba-mira.com
         
HORIBA MIRA Ltd.
 Watling Street, Nuneaton
 Warwickshire, CV10 0TU, UK
     
www.horiba-mira.com
 www.vehicletesting.solutions
 www.miratechnologypark.com
     

 If you are visiting one of our sites please use our visitor information guide for Nuneaton  and Quatro Park.
 
Attachment(s)
 image021.png
 image025.jpg
 image026.png
 image030.png
 image020.jpg
 image024.png
 image027.png
 image028.png
 image029.png
 image031.png
 image033.png
 image037.png
 image032.png
 image034.png
 image035.png
 image036.png
 2025_03_06 16_58_ROI Period 2.dts
 B7104.xml
         
 
         
   
Jordon Tray
 
Mar 7, 2025, 13:41 GMT
 
Hi Kate,
 
 
 
Ive attached the files of the other test that went wrong. I will follow up with another email containing one that was okay.
 
 
 
Regards,
 
 
     
Jordan Tray
 Engineering Technician
     
         
   
   
   
   
   
             
Direct Tel:  
   
+44 24 7635 5000
     
Email:  
   
jordan.tray@horiba-mira.com
         
HORIBA MIRA Ltd.
 Watling Street, Nuneaton
 Warwickshire, CV10 0TU, UK
     
www.horiba-mira.com
 www.vehicletesting.solutions
 www.miratechnologypark.com
     

 If you are visiting one of our sites please use our visitor information guide for Nuneaton  and Quatro Park.
 
Attachment(s)
 image020.jpg
 image021.png
 image024.png
 image025.jpg
 image026.png
 image027.png
 image028.png
 image029.png
 image030.png
 image031.png
 image032.png
 image033.png
 image034.png
 image035.png
 image036.png
 image037.png
 B7106_2025_03_04 18_14_ROI Period 2.dts
 B7106.xml
         
 
         
   
Kate Martin (DTS)
 
Mar 7, 2025, 12:18 GMT
     
Private note
     
Spoke with Jordan
  
 He is posting another test that failed and 1 practically identical that worked so we can run a delta.
         
 
         
   
Jordon Tray
 
Mar 7, 2025, 11:57 GMT
 
Hi Kate,
 
 
 
Another instance where it has happened – same deal…
 
 
 
 
 
     
Jordan Tray
 Engineering Technician
     
         
   
   
   
   
   
             
Direct Tel:  
   
+44 24 7635 5000
     
Email:  
   
jordan.tray@horiba-mira.com
         
HORIBA MIRA Ltd.
 Watling Street, Nuneaton
 Warwickshire, CV10 0TU, UK
     
www.horiba-mira.com
 www.vehicletesting.solutions
 www.miratechnologypark.com
     

 If you are visiting one of our sites please use our visitor information guide for Nuneaton  and Quatro Park.
 
Attachment(s)
 image021.jpg
 image025.png
 image026.jpg
 image028.png
 image022.png
 image027.png
 image029.png
 image030.png
 image033.png
 image035.png
 image031.png
 image032.png
 image034.png
 image036.png
 image037.png
 image038.png
         
 
         
   
Jordon Tray
 
Mar 7, 2025, 11:52 GMT
 
Hi Kate,
 
 
 
I am also stumped, a test ran yesterday and was fine..
 
 
 
Just cant work out the whole 45.0619 seconds time scale.
 
 
     
Jordan Tray
 Engineering Technician
     
         
   
   
   
   
   
             
Direct Tel:  
   
+44 24 7635 5000
     
Email:  
   
jordan.tray@horiba-mira.com
         
HORIBA MIRA Ltd.
 Watling Street, Nuneaton
 Warwickshire, CV10 0TU, UK
     
www.horiba-mira.com
 www.vehicletesting.solutions
 www.miratechnologypark.com
     

 If you are visiting one of our sites please use our visitor information guide for Nuneaton  and Quatro Park.
 
Attachment(s)
 image025.jpg
 image026.png
 image028.png
 image020.jpg
 image021.png
 image024.png
 image027.png
 image031.png
 image033.png
 image035.png
 image036.png
 image037.png
 image029.png
 image030.png
 image032.png
 image034.png
         
 
         
   
Kate Martin (DTS)
 
Mar 7, 2025, 11:19 GMT
 
 
  
 Hi Jordan
  
 I've just run through this and all good on the ROI windows when downloaded.
  
 
  
  
 I'm a bit stumped.
  
 SPS00247 look just fine compared to the other SPSs in the test for total recording and trigger samples ...
  
 
  
         
 
         
   
Jordon Tray
 
Mar 7, 2025, 10:36 GMT
 
45.0619 seems an awfully specific time to go for.
 
 
 
Is it possible it’s a memory or file serialization issue when it is being picked up the data is corrupting and changing the value?
 
 
 
I have located the SPS in question from the post test photos..
 
 
     
Jordan Tray
 Engineering Technician
     
         
   
   
   
   
   
             
Direct Tel:  
   
+44 24 7635 5000
     
Email:  
   
jordan.tray@horiba-mira.com
         
HORIBA MIRA Ltd.
 Watling Street, Nuneaton
 Warwickshire, CV10 0TU, UK
     
www.horiba-mira.com
 www.vehicletesting.solutions
 www.miratechnologypark.com
     

 If you are visiting one of our sites please use our visitor information guide for Nuneaton  and Quatro Park.
 
Attachment(s)
 image008.jpg
 image009.jpg
 image010.png
 image020.png
 image021.png
 image022.png
 image023.png
 image011.png
 image029.png
 image024.png
 image025.png
 image026.png
 image027.png
 image028.png
         
 
         
   
Kate Martin (DTS)
 
Mar 7, 2025, 09:40 GMT
 
Yes I agree - you asked for 65 secs in the test set up.
  
 The only place this can be changed is in the download section you can override the presets with a new value before you hit run but why would someone do that.
  
 I'm going to gather some DAS together and see if I can replicate it....
         
 
         
   
Jordon Tray
 
Mar 7, 2025, 09:28 GMT
 
 
 
     
Jordan Tray
 Engineering Technician
     
         
   
   
   
   
   
             
Direct Tel:  
   
+44 24 7635 5000
     
Email:  
   
jordan.tray@horiba-mira.com
         
HORIBA MIRA Ltd.
 Watling Street, Nuneaton
 Warwickshire, CV10 0TU, UK
     
www.horiba-mira.com
 www.vehicletesting.solutions
 www.miratechnologypark.com
     

 If you are visiting one of our sites please use our visitor information guide for Nuneaton  and Quatro Park.
 
Attachment(s)
 image021.jpg
 image025.png
 image027.png
 image022.jpg
 image023.png
 image024.png
 image026.png
 image029.png
 image030.png
 image031.png
 image032.png
 image028.png
         
 
         
   
Jordon Tray
 
Mar 7, 2025, 09:21 GMT
 
Hi Kate,
 
 
 
That is bizarre – If you check the test setup it is set to 65?
 
 
 
Regards,
 
 
     
Jordan Tray
 Engineering Technician
     
         
   
   
   
   
   
             
Direct Tel:  
   
+44 24 7635 5000
     
Email:  
   
jordan.tray@horiba-mira.com
         
HORIBA MIRA Ltd.
 Watling Street, Nuneaton
 Warwickshire, CV10 0TU, UK
     
www.horiba-mira.com
 www.vehicletesting.solutions
 www.miratechnologypark.com
     

 If you are visiting one of our sites please use our visitor information guide for Nuneaton  and Quatro Park.
 
Attachment(s)
 image020.jpg
 image021.jpg
 image023.png
 image022.png
 image025.png
 image028.png
 image029.png
 image031.png
 image024.png
 image026.png
 image027.png
 image030.png
         
 
         
   
Kate Martin (DTS)
 
Mar 7, 2025, 09:03 GMT
 
Thank you Jordan.
  
 This is interesting in the .dts header for the Period 2 binary files :-
  
 &lt;Module AaFilterRateHz="4000" Number="0" SerialNumber="SPS00247" NumberOfSamples="1001239" UnsubsampledNumberOfSamples="0" RequestedPostTriggerSeconds="45.0619" RequestedPreTriggerSeconds="-5" PostTriggerSeconds="90" PreTriggerSeconds="0" RecordingMode="HybridRecorderMode" SampleRateHz="20000" StartRecordSampleNumber="1439259" NumberOfChannels="3" InlineSerializedData="False" BaseSerialNumber="SPS00247" StartRecordTimestampSec="0" StartRecordTimestampNanoSec="0" TriggerTimestampSec="0" TriggerTimestampNanoSec="0" PTPMasterSync="False" TiltSensorAxisXDegreesPre="NaN" TiltSensorAxisYDegreesPre="NaN" TiltSensorAxisZDegreesPre="NaN" TiltSensorAxisXDegreesPost="NaN" TiltSensorAxisYDegreesPost="NaN" TiltSensorAxisZDegreesPost="NaN" TemperatureLocation1Pre="NaN" TemperatureLocation2Pre="NaN" TemperatureLocation3Pre="NaN" TemperatureLocation4Pre="NaN" TemperatureLocation1Post="NaN" TemperatureLocation2Post="NaN" TemperatureLocation3Post="NaN" TemperatureLocation4Post="NaN" SystemID="" SystemLocation="" TargetAxisX="NaN" TargetAxisY="NaN" TargetAxisZ="NaN" InputVoltage="NaN" BatteryVoltage="NaN"&gt;
  
 So it looks like the SW is doing exactly as requested but now I need to work out where that 45.0619 is coming from....
  
 Thxs
  
  
  
  
  
  
  
  
  
  
         
 
         
   
Jordon Tray
 
Mar 7, 2025, 08:39 GMT
 
Hi Kate,
 
 
 
Just to be clear the 45 secs that you refer to is when viewing the raw binary data in DataPro or only the exported ISO MME data in ROI - download ALL is not effected ?
 
 
 
That’s correct yes.
 
 
 
I have attached the binary &amp; exports for the ROI.
 
 
 
Regards,
 
 
     
Jordan Tray
 Engineering Technician
     
         
   
   
   
   
   
             
Direct Tel:  
   
+44 24 7635 5000
     
Email:  
   
jordan.tray@horiba-mira.com
         
HORIBA MIRA Ltd.
 Watling Street, Nuneaton
 Warwickshire, CV10 0TU, UK
     
www.horiba-mira.com
 www.vehicletesting.solutions
 www.miratechnologypark.com
     

 If you are visiting one of our sites please use our visitor information guide for Nuneaton  and Quatro Park.
 
Attachment(s)
 image017.jpg
 image018.png
 image015.jpg
 image019.png
 image020.png
 image022.png
 image023.png
 b7455binaryexportsROI.7z
 image021.png
         
 
         
   
Kate Martin (DTS)
 
Mar 6, 2025, 20:05 GMT
 
Hi Jordan
  
 We are still looking into this sorry.
  
 Just to be clear the 45 secs that you refer to is when viewing the raw binary data in DataPro or only the exported ISO MME data in ROI - download ALL is not effected ?
  
 Can you perhaps share the binary files please ?
  
 Thank you 
 Kate
  
         
 
         
   
Kate Martin (DTS)
 
Mar 6, 2025, 18:25 GMT
     
Private note
     
Thxs Dan - exactly the 2nd ROI should be 65 secs but they are reporting only 45 secs. Any download error wouldn't allow the user to view "just" 45 secs of data as the CRC would be wrong etc so I'm not sure where that is coming in. I will ask for the full data set as maybe the problem isn;t in the actual binary but the export files ? I made an unverified assumption they were taling about binary so will clarify
  
 
         
 
         
   
Dan McFadden (DTS)
 
Mar 6, 2025, 18:08 GMT
     
Private note
     
Looking at the setup I see both an original form of the ROI and the newer forms of ROI
 
 
 
        &lt;RegionsOfInterest&gt;
 
          &lt;RegionOfInterest&gt;
 
            &lt;Suffix&gt;_ROI Period 1&lt;/Suffix&gt;
 
            &lt;Start&gt;-0.1&lt;/Start&gt;
 
            &lt;End&gt;0.4&lt;/End&gt;
 
            &lt;ChannelNames&gt;\053935,\052629,\048328,\053685,\054160,\054159,\053613,\052740,\053929,\048839,\053984,\051834,\053914,\053657,\053912,\053689,\050822,\051868,\051080,\053957,\054413,\054181,\051871,\053178,\049622,\049755,\053675,\039886,\038688,\039891,\049117,\039884,\031918,\054052,\048237,\049906,\044478,\044479,\044480,\044493,\044494,\044495,\052062,\052063,\052064,\044463,\044464,\044465,\044466,\044467,\044468,\044469,\044470,\044471,\044472,\044473,\044474,\052067,\052068,\052069,\044484,\044485,\044486,\044487,\044488,\044489,\052061,\052065,\052066,\044459,\044460,\043761,\043840,\043043,\043624,[BR52226] CH-02\051105,[BR52226] CH-01\051106,[BR52443] CH-03\051107,[BR52226] CH-03\043721,[BR52443] CH-01\043722,[BR52443] CH-02\043723,[BR52233] CH-01\043725,[BR52233] CH-02\043726,[BR52233] CH-03\043727,[BR52251] CH-01\043728,[BR52251] CH-02\043729,[BR52251] CH-03\043730,[BR52431] CH-01\043731,[BR52431] CH-02\043732,[BR52431] CH-03\043733,[BR52273] CH-01\043734,[BR52273] CH-02\043735,[BR52273] CH-03\043736,[BR52230] CH-01\051114,[BR52230] CH-03\051112,[BR52230] CH-02\051113,\043743,\043741,\043742,[BR52460] CH-03\043048,[BR52460] CH-01\043635,[BR52421] CH-01\043752,[BR52421] CH-02\043753,[BR52421] CH-03\043754,[BR52428] CH-01\043755,[BR52428] CH-02\043756,[BR52428] CH-03\043757,[BR52298] CH-03\043764,[BR52288] CH-03\043763,[BR52288] CH-01\043762,[BR52228] CH-01\044461,\051110,[BR52228] CH-03\051109,\051108,\051111,\043747,\043748,\043749,\043744,\043745,\043746,\045610,\044948,\050859,\045612,\044295,\045043,\037987,\045139,\037521,\051840,\052760,\052672,\051002,\053704,\052631,[BR52069] CH-03\053396,[BR52069] CH-02\053395,[BR52069] CH-01\053394,[BR52172] CH-03\047427,\042364,\047429,[BR52929] CH-03\048167,[BR52929] CH-02\048166,[BR52929] CH-01\048165,[BR52049] CH-03\046319,[BR52099] CH-03\040829,[BR52086] CH-02\040895,[BR52063] CH-02\040902,[BR52086] CH-03\046317,[BR52049] CH-02\040890,[BR52080] CH-01\050097,[BR52080] CH-03\050096,[BR52080] CH-02\050098,[BR52928] CH-01\040852,[BR52928] CH-02\040853,[BR52928] CH-03\040855,[BR52281] CH-01\040856,[BR52281] CH-02\040857,[BR52281] CH-03\040858,[BR52104] CH-01\040866,[BR52104] CH-02\040867,[BR52104] CH-03\040868,[BR52070] CH-03\051797,[BR52070] CH-02\051796,[BR52070] CH-01\051795,[BR52041] CH-01\040875,[BR52041] CH-02\040878,[BR52041] CH-03\040880,[BR52100] CH-01\040881,[BR52100] CH-02\040882,[BR52100] CH-03\040885,[BR52050] CH-03\042901,[BR52050] CH-02\042900,[BR52050] CH-01\042899,[BR52033] CH-01\040859,[BR52049] CH-01\040849,[BR52063] CH-03\040913,[BR52063] CH-01\040911,[BR52099] CH-01\043081,[BR52099] CH-02\046318,[BR52086] CH-01\046320,[BR52159] CH-01\040869,[BR52159] CH-02\040872,[BR52159] CH-03\040874,[BR52053] CH-01\040863,[BR52053] CH-02\040864,[BR52053] CH-03\040865,[BR52057] CH-01\040860,[BR52057] CH-02\040861,[BR52057] CH-03\040862&lt;/ChannelNames&gt;
 
          &lt;/RegionOfInterest&gt;
 
          &lt;RegionOfInterest&gt;
 
            &lt;Suffix&gt;_ROI Period 2&lt;/Suffix&gt;
 
            &lt;Start&gt;-5&lt;/Start&gt;
 
            &lt;End&gt;65&lt;/End&gt;
 
            &lt;ChannelNames&gt;\043275,\043283,\043270&lt;/ChannelNames&gt;
 
          &lt;/RegionOfInterest&gt;
 
        &lt;/RegionsOfInterest&gt;
 
        &lt;ROIStart&gt;-0.1&lt;/ROIStart&gt;
 
        &lt;ROIEnd&gt;0.4&lt;/ROIEnd&gt;
 
 
 
\043275 \043283 and \043270 all appear to be in ROI 2 which should go -5 to 65s, according to kate’s screenshot there should be 153.9259s available post trigger [(3,341,425-1,539,259)/10,000]
 
 
 
Looking at the code I’m guessing this should be hardwarechannelname\\somevalue like:
 
                chanName = hardwareChannelName + "\\" + originalChannelName;
 
so I’m assuming these channels were not assigned hardware channels at test setup time and are maybe EID or manually assigned at runtime
 
 
 
043275 matches &lt;Comment&gt;EV Monitor - V2 - 1V2&lt;/Comment&gt;
 
043283 matches &lt;UserValue1&gt;EV Monitor - VB - 1VB&lt;/UserValue1&gt;
 
043270 matches &lt;UserValue1&gt;EV Monitor - V1 - 1V1&lt;/UserValue1&gt;
 
 
 
So I would naively expect these 3 channels to have the -5s to 65s roi period 2
 
 
 
Looking at log I see DataPRO v4.2.479.  I don’t see anything relevant in terms of issues fixes since then, so I think this version is fine WRT to downloading ROI.
 
 
 
Sorry if I didn’t add anything useful.  Do we have the roi dataset?
 
 
 
 
 
 
         
 
         
   
Kate Martin (DTS)
 
Mar 6, 2025, 17:34 GMT
     
Private note
     
Just looking at the SPS in the test ...
  
 
  
 SPS00247 in line with the other units.
  
 Contiuing to look for a later problem...
         
 
         
   
Dan McFadden (DTS)
 
Mar 6, 2025, 16:53 GMT
     
Private note
     
I know the feeling :/ =)
 
 
 
Cheers
 
 
                         
                         
       
       
                                           
Dan​​​​
   
 
   
Mcfadden
             
Software Engineering Manager
                     
Diversified Technical Systems, Inc.
                     
e:
       
Dan.Mcfadden@dtsweb.com
       
|
           
o:
       
+1 562 493 0158 x137
       
|
       
www.dtsweb.com
                                         
CONFIDENTIALITY: This email and any attachments may contain information from Diversified Technical Systems, Inc. (DTS) which​ is confidential ​and/or proprietary. If you received this transmission in error, any use, reproduction or dissemination of this transmission​ is strictly prohibited. ​Please delete this message and notify sender immediately.  
 ​EXPORT CONTROL NOTICE: This e-mail may contain technical data whose export, transfer, and/or disclosure may be controlled by​ the US ​International Traffic in Arms Regulation (ITAR) 22 CFR part 120-130 or the Export Administration Regulations (Commerce).
 ​
 ​
                   
 
                 
Attachment(s)
 ~WRD3003.jpg
         
 
         
   
Kate Martin (DTS)
 
Mar 6, 2025, 16:47 GMT
     
Private note
     
Thxs Dan - that's exactly what I'm doing right now but the phone keep ringing ...
  
  
         
 
         
   
Dan McFadden (DTS)
 
Mar 6, 2025, 16:36 GMT
     
Private note
     
I would look through the logs and see how many samples are available from DAS prior to the download – something like TotalSamples – if it lost power or something maybe the full # of samples wouldn’t be available to download?
 
 
                         
                         
       
       
                                           
Dan​​​​
   
 
   
Mcfadden
             
Software Engineering Manager
                     
Diversified Technical Systems, Inc.
                     
e:
       
Dan.Mcfadden@dtsweb.com
       
|
           
o:
       
+1 562 493 0158 x137
       
|
       
www.dtsweb.com
                                         
CONFIDENTIALITY: This email and any attachments may contain information from Diversified Technical Systems, Inc. (DTS) which​ is confidential ​and/or proprietary. If you received this transmission in error, any use, reproduction or dissemination of this transmission​ is strictly prohibited. ​Please delete this message and notify sender immediately.  
 ​EXPORT CONTROL NOTICE: This e-mail may contain technical data whose export, transfer, and/or disclosure may be controlled by​ the US ​International Traffic in Arms Regulation (ITAR) 22 CFR part 120-130 or the Export Administration Regulations (Commerce).
 ​
 ​
                   
 
                         
 
         
   
Dan McFadden (DTS)
 
Mar 6, 2025, 16:34 GMT
     
Private note
     
Hi Kate,
 
 
 
That line is coming from here:
 
 
 
//30129 Cannot download ROI if Test Setup has TSR AIR and multiple ROIs
 
//34350 Don't exclude Voltage Input channels when downloading multiple ROIs
 
channel.IsConfigured = dlInfo.ChannelIdList.ToList().Contains(Convert.ToInt64(aic.ChannelId));
 
APILogger.Log($"aic: Setting channel is configured to {channel.IsConfigured} because {aic.ChannelId} was comparted to {channelIdsString} ");
 
 
 
It is determining that a channel is not configured because the list of channels to download doesn’t contain that channel – as if the ROI doesn’t include it
 
 
 
The MS tickets attached to that log are:
 
34350: Issues using voltage input channels with multiple rois and the download tab. - Manuscript
 
 
 
30129: Cannot download ROI if Test Setup has TSR AIR and multiple ROIs - Manuscript
 
 
 
 
 
I didn’t write the code in question but it was to allow downloading with multiple ROIs with channels duplicate sensor IDs that come from using voltage input channels on multiple channels, or the embedded channels in a TSR AIR.
 
 
 
 
 
 
                         
                         
       
       
                                           
Dan​​​​
   
 
   
Mcfadden
             
Software Engineering Manager
                     
Diversified Technical Systems, Inc.
                     
e:
       
Dan.Mcfadden@dtsweb.com
       
|
           
o:
       
+1 562 493 0158 x137
       
|
       
www.dtsweb.com
                                         
CONFIDENTIALITY: This email and any attachments may contain information from Diversified Technical Systems, Inc. (DTS) which​ is confidential ​and/or proprietary. If you received this transmission in error, any use, reproduction or dissemination of this transmission​ is strictly prohibited. ​Please delete this message and notify sender immediately.  
 ​EXPORT CONTROL NOTICE: This e-mail may contain technical data whose export, transfer, and/or disclosure may be controlled by​ the US ​International Traffic in Arms Regulation (ITAR) 22 CFR part 120-130 or the Export Administration Regulations (Commerce).
 ​
 ​
                   
 
                         
 
         
   
Kate Martin (DTS)
 
Mar 6, 2025, 16:18 GMT
     
Private note
     
Any idea what this means please ?
  
 2025-03-05 18:20:40.401 aic: Setting channel is configured to False because [SPS00247] CH-01\043270 was compared to [SPS00398] CH-03\053935, [SPS00398] CH-08\052629, [SPS00398] CH-11\048328, [SPS00398] CH-04\053685, [SPS00423] CH-09\054160, [SPS00423] CH-18\054159, [SPS00423] CH-14\053613, [SPS00423] CH-16\052740, [SPS00423] CH-15\053929, [SPS00398] CH-02\048839, [SPS00423] CH-17\053984, [SPS00398] CH-09\051834, [SPS00423] CH-11\053914, [SPS00423] CH-12\053657, [SPS00423] CH-13\053912, [SPS00423] CH-10\053689, [SPS00398] CH-06\050822, [SPS00398] CH-01\051868, [SPS00398] CH-10\051080, [SPS00423] CH-07\053957, [SPS00398] CH-05\054413, [SPS00423] CH-08\054181, [SPS00398] CH-07\051871, [SPS00423] CH-03\053178, [SPS00423] CH-04\049622, [SPS00423] CH-06\049755, [SPS00423] CH-02\053675, [SPS00398] CH-18\039886, [SPS00398] CH-12\038688, [SPS00398] CH-16\039891, [SPS00398] CH-15\049117, [SPS00398] CH-17\039884, [SPS00398] CH-14\031918, [SPS00398] CH-13\054052, [SPS00423]
  
 as it's SPS00247 that has the HV channels in question...
  
 
  
         
 
         
   
Jordon Tray
 
Mar 6, 2025, 11:46 GMT
 
Also the impact was at approx. 18:13.
 
 
 
Regards,
 
 
     
Jordan Tray
 Engineering Technician
     
         
   
   
   
   
   
             
Direct Tel:  
   
+44 24 7635 5000
     
Email:  
   
jordan.tray@horiba-mira.com
         
HORIBA MIRA Ltd.
 Watling Street, Nuneaton
 Warwickshire, CV10 0TU, UK
     
www.horiba-mira.com
 www.vehicletesting.solutions
 www.miratechnologypark.com
     

 If you are visiting one of our sites please use our visitor information guide for Nuneaton  and Quatro Park.
 
Attachment(s)
 image016.jpg
 image017.jpg
         
 
         
   
Jordon Tray
 
Mar 6, 2025, 11:26 GMT
 
Attached as much as possible from yesterday including the setup. I have just been informed this has happened previously on more than 1 test.
 
 
 
Regards,
 
 
     
Jordan Tray
 Engineering Technician
     
         
   
   
   
   
   
             
Direct Tel:  
   
+44 24 7635 5000
     
Email:  
   
jordan.tray@horiba-mira.com
         
HORIBA MIRA Ltd.
 Watling Street, Nuneaton
 Warwickshire, CV10 0TU, UK
     
www.horiba-mira.com
 www.vehicletesting.solutions
 www.miratechnologypark.com
     

 If you are visiting one of our sites please use our visitor information guide for Nuneaton  and Quatro Park.
 
Attachment(s)
 image016.jpg
 image017.jpg
 DASConfigs 2025-03-05 17.45.03.log.bz2
 DASConfigs 2025-03-05 18.14.36.log.bz2
 DASConfigs 2025-03-05 18.17.56.log.bz2
 DASConfigs 2025-03-05 18.20.37.log.bz2
 DASConfigs 2025-03-05 18.26.26.log.bz2
 DASConfigs.log
 DataPRO 2025-03-05 17.37.50.log.bz2
 DASConfigs 2025-03-05 18.16.50.log.bz2
 DataPRO 2025-03-05 18.12.34.log.bz2
 DataPRO 2025-03-05 18.14.30.log.bz2
 DataPRO 2025-03-05 18.18.52.log.bz2
 DataPRO 2025-03-05 18.11.17.log.bz2
 DataPRO 2025-03-05 18.13.32.log.bz2
 DataPRO 2025-03-05 18.26.41.log.bz2
 DataPRO 2025-03-05 18.27.26.log.bz2
 DataPRO 2025-03-05 18.28.10.log.bz2
 DataPRO 2025-03-05 18.21.20.log.bz2
 DataPRO 2025-03-05 18.28.56.log.bz2
 DataPRO 2025-03-05 18.29.40.log.bz2
 DataPRO 2025-03-05 18.30.27.log.bz2
 DataPRO 2025-03-05 18.32.22.log.bz2
 DataPRO 2025-03-05 19.09.32.log.bz2
 DB 2025-03-05 17.38.17.log.bz2
 DataPRO 2025-03-05 18.37.54.log.bz2
 ping.log
 DASConfigs 2025-03-05 17.41.06.log.bz2
 B7455.xml
         
 
         
   
Kate Martin (DTS)
 
Mar 6, 2025, 10:00 GMT
 
Hi Jordon
 
 Can you please attach the test set up and logs ?
 
 Thxs
         
 
         
   
Jordon Tray
 
Mar 6, 2025, 09:46 GMT
 
We have an issue where our HV ROI data from test is cutting off at 45s not 65s as set in the test setup.
         
----------------------------------------------
 Customer Details:
 Name: Jordon Tray ( jordan.tray@horiba-mira.com )
 Organization: Horiba MIRA Ltd
 
----------------------------------------------
 If you no longer wish to see notifications for this ticket, please use the 'Unfollow' option in the Followers List.
 
You are an agent. Add a comment by replying to this email or view ticket in Zendesk Support.
     
Ticket #
   
14994
     
Status
   
On-hold
     
Requester
   
Jordon Tray
     
CCs
   
-
     
Followers
   
Dan McFadden, Jerry Lawrence, Maugan Robert
     
Group
   
Support
     
Assignee
   
Kate Martin
     
Priority
   
Normal
     
Type
   
Problem
     
Channel
   
Web Form
     
 
 
This email is a service from DTS. Delivered by Zendesk.
 
[070XX0-K5L7K]Ticket-Id:14994Account-Subdomain:dtsweb
 
 
     
VPG Security WARNING: This is an external email. Do not click links or open attachments unless you recognize the sender and know the content is safe.
     

  
 
##- Please type your reply above this line -##
 
You are a follower on this request (#14994, #dtsweb.zendesk.com/agent/tickets/14994). Reply to this email to add an internal note to the request.
         
   
Kate Martin (DTS)
 
Mar 31, 2025, 15:24 GMT+1
 
Interestingly 4.2.526 appears to not have this issue ..or at least I can't replicate my theory in this version.
  
 
  
         
 
         
   
Kate Martin (DTS)
 
Mar 31, 2025, 14:08 GMT+1
 
Hi Dan, Jerry
  
 Thank you for the other "good" and "bad" tests Jordan
  
 I think I have finally twigged what is happening here - not sure why I missed it last week sorry.
  
 Here is my 90secs post T0 hybrid test set up with ROI 2 set to 65 secs...
  
  
  
 
  
  
 The key to replicating Dan is to have the DAS that is being used in ROI_2 sampling at a higher rate than the DAS in ROI_1.
  
 That was what was the delta in the 2 MIRA tests set ups - one had ATD's sampling at 10,000sps and the SPS at 20.000sps and ther "good" one had everything at 20,000sps.
  
 With the ROI_2 DAS sampling at equal to or lower than the main ROI_1 DAS all is well..
  
 
  
 But switch it around like I have here with SPS00254 (the ROI_2 DAS) sampling at 20,000 
  
  
 
  
 and even thought the logs files correctly report "total samples recorded" and "trigger sample number" for example, it is the GUI that for some reason recomputes the ROI for the DAS in ROI_2 for some reason ..
  
 I suspect maybe it is using the "total samples recorded" from the ROI_1 DAS and appling the ROI_2 DAS sample rate to that number to come up with the highly suspect 45 seconds which is 90 secs total test duration divided by "DAS ROI 2 sample rate". 
  
 Does the computation for adjusting the available ROI respect the sampling rates for all DAS in that ROI ?
  
 
  
  
  
  
 
  
 logs and test set up folder attached Dan.
  
 Thxs
  
 
Attachment(s)
 DataPRO 2025-03-31 13.49.48.log.bz2
 DataPRO 2025-03-31 13.50.24.log.bz2
 2025_03_31 13_44.7z
         
 
         
   
Kate Martin (DTS)
 
Mar 31, 2025, 12:29 GMT+1
     
Private note
     
Conclusion
  
 Good Test B7104 has all DAS at 20,000sps
  
 Bad Test B7106 has mixed DAS at 10,000sps and 20,000sps
  
  
         
 
         
   
Kate Martin (DTS)
 
Mar 31, 2025, 12:16 GMT+1
     
Private note
     
.dts file
  
 
  
  
         
 
         
   
Kate Martin (DTS)
 
Mar 31, 2025, 11:56 GMT+1
     
Private note
     
.xml files that is 
         
 
         
   
Kate Martin (DTS)
 
Mar 31, 2025, 11:55 GMT+1
     
Private note
     
Only ROI Period 2 references in .dts files look good..
  
 Good Test B7104
  
 
  
 "Bad Test"
  
 
  
  
         
 
         
   
Jordon Tray
 
Mar 17, 2025, 08:21 GMT
 
Good Morning Kate,
 
 
 
Any advancements on this? Colin keen to work out the issue as we have many EuroNCAP tests in the pipeline.
 
 
 
Thanks,
 
 
     
Jordan Tray
 Engineering Technician
     
         
   
   
   
   
   
             
Direct Tel:  
   
+44 24 7635 5000
     
Email:  
   
jordan.tray@horiba-mira.com
         
HORIBA MIRA Ltd.
 Watling Street, Nuneaton
 Warwickshire, CV10 0TU, UK
     
www.horiba-mira.com
 www.vehicletesting.solutions
 www.miratechnologypark.com
     

 If you are visiting one of our sites please use our visitor information guide for Nuneaton  and Quatro Park.
 
Attachment(s)
 image021.png
 image020.jpg
 image024.png
 image027.png
 image028.png
 image029.png
 image032.png
 image033.png
 image034.png
 image035.png
 image036.png
 image037.png
 image025.jpg
 image026.png
 image030.png
 image031.png
         
 
         
   
Dan McFadden (DTS)
 
Mar 10, 2025, 15:20 GMT
     
Private note
     
Is it possible it somehow picked up a configuration file from an unexpected location?  Maybe it a corrupted one on the unit so it used a backup on the PC and it happened to have a backup on the pc that was old for some reason?
 
 
                         
                         
       
       
                                           
Dan​​​​
   
 
   
Mcfadden
             
Software Engineering Manager
                     
Diversified Technical Systems, Inc.
                     
e:
       
Dan.Mcfadden@dtsweb.com
       
|
           
o:
       
+1 562 493 0158 x137
       
|
       
www.dtsweb.com
                                         
CONFIDENTIALITY: This email and any attachments may contain information from Diversified Technical Systems, Inc. (DTS) which​ is confidential ​and/or proprietary. If you received this transmission in error, any use, reproduction or dissemination of this transmission​ is strictly prohibited. ​Please delete this message and notify sender immediately.  
 ​EXPORT CONTROL NOTICE: This e-mail may contain technical data whose export, transfer, and/or disclosure may be controlled by​ the US ​International Traffic in Arms Regulation (ITAR) 22 CFR part 120-130 or the Export Administration Regulations (Commerce).
 ​
 ​
                   
 
                 
Attachment(s)
 ~WRD0004.jpg
         
 
         
   
Jordon Tray
 
Mar 7, 2025, 13:48 GMT
 
Attached here is working test..
 
 
     
Jordan Tray
 Engineering Technician
     
         
   
   
   
   
   
             
Direct Tel:  
   
+44 24 7635 5000
     
Email:  
   
jordan.tray@horiba-mira.com
         
HORIBA MIRA Ltd.
 Watling Street, Nuneaton
 Warwickshire, CV10 0TU, UK
     
www.horiba-mira.com
 www.vehicletesting.solutions
 www.miratechnologypark.com
     

 If you are visiting one of our sites please use our visitor information guide for Nuneaton  and Quatro Park.
 
Attachment(s)
 image021.png
 image025.jpg
 image026.png
 image030.png
 image020.jpg
 image024.png
 image027.png
 image028.png
 image029.png
 image031.png
 image033.png
 image037.png
 image032.png
 image034.png
 image035.png
 image036.png
 2025_03_06 16_58_ROI Period 2.dts
 B7104.xml
         
 
         
   
Jordon Tray
 
Mar 7, 2025, 13:41 GMT
 
Hi Kate,
 
 
 
Ive attached the files of the other test that went wrong. I will follow up with another email containing one that was okay.
 
 
 
Regards,
 
 
     
Jordan Tray
 Engineering Technician
     
         
   
   
   
   
   
             
Direct Tel:  
   
+44 24 7635 5000
     
Email:  
   
jordan.tray@horiba-mira.com
         
HORIBA MIRA Ltd.
 Watling Street, Nuneaton
 Warwickshire, CV10 0TU, UK
     
www.horiba-mira.com
 www.vehicletesting.solutions
 www.miratechnologypark.com
     

 If you are visiting one of our sites please use our visitor information guide for Nuneaton  and Quatro Park.
 
Attachment(s)
 image020.jpg
 image021.png
 image024.png
 image025.jpg
 image026.png
 image027.png
 image028.png
 image029.png
 image030.png
 image031.png
 image032.png
 image033.png
 image034.png
 image035.png
 image036.png
 image037.png
 B7106_2025_03_04 18_14_ROI Period 2.dts
 B7106.xml
         
 
         
   
Kate Martin (DTS)
 
Mar 7, 2025, 12:18 GMT
     
Private note
     
Spoke with Jordan
  
 He is posting another test that failed and 1 practically identical that worked so we can run a delta.
         
 
         
   
Jordon Tray
 
Mar 7, 2025, 11:57 GMT
 
Hi Kate,
 
 
 
Another instance where it has happened – same deal…
 
 
 
 
 
     
Jordan Tray
 Engineering Technician
     
         
   
   
   
   
   
             
Direct Tel:  
   
+44 24 7635 5000
     
Email:  
   
jordan.tray@horiba-mira.com
         
HORIBA MIRA Ltd.
 Watling Street, Nuneaton
 Warwickshire, CV10 0TU, UK
     
www.horiba-mira.com
 www.vehicletesting.solutions
 www.miratechnologypark.com
     

 If you are visiting one of our sites please use our visitor information guide for Nuneaton  and Quatro Park.
 
Attachment(s)
 image021.jpg
 image025.png
 image026.jpg
 image028.png
 image022.png
 image027.png
 image029.png
 image030.png
 image033.png
 image035.png
 image031.png
 image032.png
 image034.png
 image036.png
 image037.png
 image038.png
         
 
         
   
Jordon Tray
 
Mar 7, 2025, 11:52 GMT
 
Hi Kate,
 
 
 
I am also stumped, a test ran yesterday and was fine..
 
 
 
Just cant work out the whole 45.0619 seconds time scale.
 
 
     
Jordan Tray
 Engineering Technician
     
         
   
   
   
   
   
             
Direct Tel:  
   
+44 24 7635 5000
     
Email:  
   
jordan.tray@horiba-mira.com
         
HORIBA MIRA Ltd.
 Watling Street, Nuneaton
 Warwickshire, CV10 0TU, UK
     
www.horiba-mira.com
 www.vehicletesting.solutions
 www.miratechnologypark.com
     

 If you are visiting one of our sites please use our visitor information guide for Nuneaton  and Quatro Park.
 
Attachment(s)
 image025.jpg
 image026.png
 image028.png
 image020.jpg
 image021.png
 image024.png
 image027.png
 image031.png
 image033.png
 image035.png
 image036.png
 image037.png
 image029.png
 image030.png
 image032.png
 image034.png
         
 
         
   
Kate Martin (DTS)
 
Mar 7, 2025, 11:19 GMT
 
 
  
 Hi Jordan
  
 I've just run through this and all good on the ROI windows when downloaded.
  
 
  
  
 I'm a bit stumped.
  
 SPS00247 look just fine compared to the other SPSs in the test for total recording and trigger samples ...
  
 
  
         
 
         
   
Jordon Tray
 
Mar 7, 2025, 10:36 GMT
 
45.0619 seems an awfully specific time to go for.
 
 
 
Is it possible it’s a memory or file serialization issue when it is being picked up the data is corrupting and changing the value?
 
 
 
I have located the SPS in question from the post test photos..
 
 
     
Jordan Tray
 Engineering Technician
     
         
   
   
   
   
   
             
Direct Tel:  
   
+44 24 7635 5000
     
Email:  
   
jordan.tray@horiba-mira.com
         
HORIBA MIRA Ltd.
 Watling Street, Nuneaton
 Warwickshire, CV10 0TU, UK
     
www.horiba-mira.com
 www.vehicletesting.solutions
 www.miratechnologypark.com
     

 If you are visiting one of our sites please use our visitor information guide for Nuneaton  and Quatro Park.
 
Attachment(s)
 image008.jpg
 image009.jpg
 image010.png
 image020.png
 image021.png
 image022.png
 image023.png
 image011.png
 image029.png
 image024.png
 image025.png
 image026.png
 image027.png
 image028.png
         
 
         
   
Kate Martin (DTS)
 
Mar 7, 2025, 09:40 GMT
 
Yes I agree - you asked for 65 secs in the test set up.
  
 The only place this can be changed is in the download section you can override the presets with a new value before you hit run but why would someone do that.
  
 I'm going to gather some DAS together and see if I can replicate it....
         
 
         
   
Jordon Tray
 
Mar 7, 2025, 09:28 GMT
 
 
 
     
Jordan Tray
 Engineering Technician
     
         
   
   
   
   
   
             
Direct Tel:  
   
+44 24 7635 5000
     
Email:  
   
jordan.tray@horiba-mira.com
         
HORIBA MIRA Ltd.
 Watling Street, Nuneaton
 Warwickshire, CV10 0TU, UK
     
www.horiba-mira.com
 www.vehicletesting.solutions
 www.miratechnologypark.com
     

 If you are visiting one of our sites please use our visitor information guide for Nuneaton  and Quatro Park.
 
Attachment(s)
 image021.jpg
 image025.png
 image027.png
 image022.jpg
 image023.png
 image024.png
 image026.png
 image029.png
 image030.png
 image031.png
 image032.png
 image028.png
         
 
         
   
Jordon Tray
 
Mar 7, 2025, 09:21 GMT
 
Hi Kate,
 
 
 
That is bizarre – If you check the test setup it is set to 65?
 
 
 
Regards,
 
 
     
Jordan Tray
 Engineering Technician
     
         
   
   
   
   
   
             
Direct Tel:  
   
+44 24 7635 5000
     
Email:  
   
jordan.tray@horiba-mira.com
         
HORIBA MIRA Ltd.
 Watling Street, Nuneaton
 Warwickshire, CV10 0TU, UK
     
www.horiba-mira.com
 www.vehicletesting.solutions
 www.miratechnologypark.com
     

 If you are visiting one of our sites please use our visitor information guide for Nuneaton  and Quatro Park.
 
Attachment(s)
 image020.jpg
 image021.jpg
 image023.png
 image022.png
 image025.png
 image028.png
 image029.png
 image031.png
 image024.png
 image026.png
 image027.png
 image030.png
         
 
         
   
Kate Martin (DTS)
 
Mar 7, 2025, 09:03 GMT
 
Thank you Jordan.
  
 This is interesting in the .dts header for the Period 2 binary files :-
  
 &lt;Module AaFilterRateHz="4000" Number="0" SerialNumber="SPS00247" NumberOfSamples="1001239" UnsubsampledNumberOfSamples="0" RequestedPostTriggerSeconds="45.0619" RequestedPreTriggerSeconds="-5" PostTriggerSeconds="90" PreTriggerSeconds="0" RecordingMode="HybridRecorderMode" SampleRateHz="20000" StartRecordSampleNumber="1439259" NumberOfChannels="3" InlineSerializedData="False" BaseSerialNumber="SPS00247" StartRecordTimestampSec="0" StartRecordTimestampNanoSec="0" TriggerTimestampSec="0" TriggerTimestampNanoSec="0" PTPMasterSync="False" TiltSensorAxisXDegreesPre="NaN" TiltSensorAxisYDegreesPre="NaN" TiltSensorAxisZDegreesPre="NaN" TiltSensorAxisXDegreesPost="NaN" TiltSensorAxisYDegreesPost="NaN" TiltSensorAxisZDegreesPost="NaN" TemperatureLocation1Pre="NaN" TemperatureLocation2Pre="NaN" TemperatureLocation3Pre="NaN" TemperatureLocation4Pre="NaN" TemperatureLocation1Post="NaN" TemperatureLocation2Post="NaN" TemperatureLocation3Post="NaN" TemperatureLocation4Post="NaN" SystemID="" SystemLocation="" TargetAxisX="NaN" TargetAxisY="NaN" TargetAxisZ="NaN" InputVoltage="NaN" BatteryVoltage="NaN"&gt;
  
 So it looks like the SW is doing exactly as requested but now I need to work out where that 45.0619 is coming from....
  
 Thxs
  
  
  
  
  
  
  
  
  
  
         
 
         
   
Jordon Tray
 
Mar 7, 2025, 08:39 GMT
 
Hi Kate,
 
 
 
Just to be clear the 45 secs that you refer to is when viewing the raw binary data in DataPro or only the exported ISO MME data in ROI - download ALL is not effected ?
 
 
 
That’s correct yes.
 
 
 
I have attached the binary &amp; exports for the ROI.
 
 
 
Regards,
 
 
     
Jordan Tray
 Engineering Technician
     
         
   
   
   
   
   
             
Direct Tel:  
   
+44 24 7635 5000
     
Email:  
   
jordan.tray@horiba-mira.com
         
HORIBA MIRA Ltd.
 Watling Street, Nuneaton
 Warwickshire, CV10 0TU, UK
     
www.horiba-mira.com
 www.vehicletesting.solutions
 www.miratechnologypark.com
     

 If you are visiting one of our sites please use our visitor information guide for Nuneaton  and Quatro Park.
 
Attachment(s)
 image017.jpg
 image018.png
 image015.jpg
 image019.png
 image020.png
 image022.png
 image023.png
 b7455binaryexportsROI.7z
 image021.png
         
 
         
   
Kate Martin (DTS)
 
Mar 6, 2025, 20:05 GMT
 
Hi Jordan
  
 We are still looking into this sorry.
  
 Just to be clear the 45 secs that you refer to is when viewing the raw binary data in DataPro or only the exported ISO MME data in ROI - download ALL is not effected ?
  
 Can you perhaps share the binary files please ?
  
 Thank you 
 Kate
  
         
 
         
   
Kate Martin (DTS)
 
Mar 6, 2025, 18:25 GMT
     
Private note
     
Thxs Dan - exactly the 2nd ROI should be 65 secs but they are reporting only 45 secs. Any download error wouldn't allow the user to view "just" 45 secs of data as the CRC would be wrong etc so I'm not sure where that is coming in. I will ask for the full data set as maybe the problem isn;t in the actual binary but the export files ? I made an unverified assumption they were taling about binary so will clarify
  
 
         
 
         
   
Dan McFadden (DTS)
 
Mar 6, 2025, 18:08 GMT
     
Private note
     
Looking at the setup I see both an original form of the ROI and the newer forms of ROI
 
 
 
        &lt;RegionsOfInterest&gt;
 
          &lt;RegionOfInterest&gt;
 
            &lt;Suffix&gt;_ROI Period 1&lt;/Suffix&gt;
 
            &lt;Start&gt;-0.1&lt;/Start&gt;
 
            &lt;End&gt;0.4&lt;/End&gt;
 
            &lt;ChannelNames&gt;\053935,\052629,\048328,\053685,\054160,\054159,\053613,\052740,\053929,\048839,\053984,\051834,\053914,\053657,\053912,\053689,\050822,\051868,\051080,\053957,\054413,\054181,\051871,\053178,\049622,\049755,\053675,\039886,\038688,\039891,\049117,\039884,\031918,\054052,\048237,\049906,\044478,\044479,\044480,\044493,\044494,\044495,\052062,\052063,\052064,\044463,\044464,\044465,\044466,\044467,\044468,\044469,\044470,\044471,\044472,\044473,\044474,\052067,\052068,\052069,\044484,\044485,\044486,\044487,\044488,\044489,\052061,\052065,\052066,\044459,\044460,\043761,\043840,\043043,\043624,[BR52226] CH-02\051105,[BR52226] CH-01\051106,[BR52443] CH-03\051107,[BR52226] CH-03\043721,[BR52443] CH-01\043722,[BR52443] CH-02\043723,[BR52233] CH-01\043725,[BR52233] CH-02\043726,[BR52233] CH-03\043727,[BR52251] CH-01\043728,[BR52251] CH-02\043729,[BR52251] CH-03\043730,[BR52431] CH-01\043731,[BR52431] CH-02\043732,[BR52431] CH-03\043733,[BR52273] CH-01\043734,[BR52273] CH-02\043735,[BR52273] CH-03\043736,[BR52230] CH-01\051114,[BR52230] CH-03\051112,[BR52230] CH-02\051113,\043743,\043741,\043742,[BR52460] CH-03\043048,[BR52460] CH-01\043635,[BR52421] CH-01\043752,[BR52421] CH-02\043753,[BR52421] CH-03\043754,[BR52428] CH-01\043755,[BR52428] CH-02\043756,[BR52428] CH-03\043757,[BR52298] CH-03\043764,[BR52288] CH-03\043763,[BR52288] CH-01\043762,[BR52228] CH-01\044461,\051110,[BR52228] CH-03\051109,\051108,\051111,\043747,\043748,\043749,\043744,\043745,\043746,\045610,\044948,\050859,\045612,\044295,\045043,\037987,\045139,\037521,\051840,\052760,\052672,\051002,\053704,\052631,[BR52069] CH-03\053396,[BR52069] CH-02\053395,[BR52069] CH-01\053394,[BR52172] CH-03\047427,\042364,\047429,[BR52929] CH-03\048167,[BR52929] CH-02\048166,[BR52929] CH-01\048165,[BR52049] CH-03\046319,[BR52099] CH-03\040829,[BR52086] CH-02\040895,[BR52063] CH-02\040902,[BR52086] CH-03\046317,[BR52049] CH-02\040890,[BR52080] CH-01\050097,[BR52080] CH-03\050096,[BR52080] CH-02\050098,[BR52928] CH-01\040852,[BR52928] CH-02\040853,[BR52928] CH-03\040855,[BR52281] CH-01\040856,[BR52281] CH-02\040857,[BR52281] CH-03\040858,[BR52104] CH-01\040866,[BR52104] CH-02\040867,[BR52104] CH-03\040868,[BR52070] CH-03\051797,[BR52070] CH-02\051796,[BR52070] CH-01\051795,[BR52041] CH-01\040875,[BR52041] CH-02\040878,[BR52041] CH-03\040880,[BR52100] CH-01\040881,[BR52100] CH-02\040882,[BR52100] CH-03\040885,[BR52050] CH-03\042901,[BR52050] CH-02\042900,[BR52050] CH-01\042899,[BR52033] CH-01\040859,[BR52049] CH-01\040849,[BR52063] CH-03\040913,[BR52063] CH-01\040911,[BR52099] CH-01\043081,[BR52099] CH-02\046318,[BR52086] CH-01\046320,[BR52159] CH-01\040869,[BR52159] CH-02\040872,[BR52159] CH-03\040874,[BR52053] CH-01\040863,[BR52053] CH-02\040864,[BR52053] CH-03\040865,[BR52057] CH-01\040860,[BR52057] CH-02\040861,[BR52057] CH-03\040862&lt;/ChannelNames&gt;
 
          &lt;/RegionOfInterest&gt;
 
          &lt;RegionOfInterest&gt;
 
            &lt;Suffix&gt;_ROI Period 2&lt;/Suffix&gt;
 
            &lt;Start&gt;-5&lt;/Start&gt;
 
            &lt;End&gt;65&lt;/End&gt;
 
            &lt;ChannelNames&gt;\043275,\043283,\043270&lt;/ChannelNames&gt;
 
          &lt;/RegionOfInterest&gt;
 
        &lt;/RegionsOfInterest&gt;
 
        &lt;ROIStart&gt;-0.1&lt;/ROIStart&gt;
 
        &lt;ROIEnd&gt;0.4&lt;/ROIEnd&gt;
 
 
 
\043275 \043283 and \043270 all appear to be in ROI 2 which should go -5 to 65s, according to kate’s screenshot there should be 153.9259s available post trigger [(3,341,425-1,539,259)/10,000]
 
 
 
Looking at the code I’m guessing this should be hardwarechannelname\\somevalue like:
 
                chanName = hardwareChannelName + "\\" + originalChannelName;
 
so I’m assuming these channels were not assigned hardware channels at test setup time and are maybe EID or manually assigned at runtime
 
 
 
043275 matches &lt;Comment&gt;EV Monitor - V2 - 1V2&lt;/Comment&gt;
 
043283 matches &lt;UserValue1&gt;EV Monitor - VB - 1VB&lt;/UserValue1&gt;
 
043270 matches &lt;UserValue1&gt;EV Monitor - V1 - 1V1&lt;/UserValue1&gt;
 
 
 
So I would naively expect these 3 channels to have the -5s to 65s roi period 2
 
 
 
Looking at log I see DataPRO v4.2.479.  I don’t see anything relevant in terms of issues fixes since then, so I think this version is fine WRT to downloading ROI.
 
 
 
Sorry if I didn’t add anything useful.  Do we have the roi dataset?
 
 
 
 
 
 
         
 
         
   
Kate Martin (DTS)
 
Mar 6, 2025, 17:34 GMT
     
Private note
     
Just looking at the SPS in the test ...
  
 
  
 SPS00247 in line with the other units.
  
 Contiuing to look for a later problem...
         
 
         
   
Dan McFadden (DTS)
 
Mar 6, 2025, 16:53 GMT
     
Private note
     
I know the feeling :/ =)
 
 
 
Cheers
 
 
                         
                         
       
       
                                           
Dan​​​​
   
 
   
Mcfadden
             
Software Engineering Manager
                     
Diversified Technical Systems, Inc.
                     
e:
       
Dan.Mcfadden@dtsweb.com
       
|
           
o:
       
+1 562 493 0158 x137
       
|
       
www.dtsweb.com
                                         
CONFIDENTIALITY: This email and any attachments may contain information from Diversified Technical Systems, Inc. (DTS) which​ is confidential ​and/or proprietary. If you received this transmission in error, any use, reproduction or dissemination of this transmission​ is strictly prohibited. ​Please delete this message and notify sender immediately.  
 ​EXPORT CONTROL NOTICE: This e-mail may contain technical data whose export, transfer, and/or disclosure may be controlled by​ the US ​International Traffic in Arms Regulation (ITAR) 22 CFR part 120-130 or the Export Administration Regulations (Commerce).
 ​
 ​
                   
 
                 
Attachment(s)
 ~WRD3003.jpg
         
 
         
   
Kate Martin (DTS)
 
Mar 6, 2025, 16:47 GMT
     
Private note
     
Thxs Dan - that's exactly what I'm doing right now but the phone keep ringing ...
  
  
         
 
         
   
Dan McFadden (DTS)
 
Mar 6, 2025, 16:36 GMT
     
Private note
     
I would look through the logs and see how many samples are available from DAS prior to the download – something like TotalSamples – if it lost power or something maybe the full # of samples wouldn’t be available to download?
 
 
                         
                         
       
       
                                           
Dan​​​​
   
 
   
Mcfadden
             
Software Engineering Manager
                     
Diversified Technical Systems, Inc.
                     
e:
       
Dan.Mcfadden@dtsweb.com
       
|
           
o:
       
+1 562 493 0158 x137
       
|
       
www.dtsweb.com
                                         
CONFIDENTIALITY: This email and any attachments may contain information from Diversified Technical Systems, Inc. (DTS) which​ is confidential ​and/or proprietary. If you received this transmission in error, any use, reproduction or dissemination of this transmission​ is strictly prohibited. ​Please delete this message and notify sender immediately.  
 ​EXPORT CONTROL NOTICE: This e-mail may contain technical data whose export, transfer, and/or disclosure may be controlled by​ the US ​International Traffic in Arms Regulation (ITAR) 22 CFR part 120-130 or the Export Administration Regulations (Commerce).
 ​
 ​
                   
 
                         
 
         
   
Dan McFadden (DTS)
 
Mar 6, 2025, 16:34 GMT
     
Private note
     
Hi Kate,
 
 
 
That line is coming from here:
 
 
 
//30129 Cannot download ROI if Test Setup has TSR AIR and multiple ROIs
 
//34350 Don't exclude Voltage Input channels when downloading multiple ROIs
 
channel.IsConfigured = dlInfo.ChannelIdList.ToList().Contains(Convert.ToInt64(aic.ChannelId));
 
APILogger.Log($"aic: Setting channel is configured to {channel.IsConfigured} because {aic.ChannelId} was comparted to {channelIdsString} ");
 
 
 
It is determining that a channel is not configured because the list of channels to download doesn’t contain that channel – as if the ROI doesn’t include it
 
 
 
The MS tickets attached to that log are:
 
34350: Issues using voltage input channels with multiple rois and the download tab. - Manuscript
 
 
 
30129: Cannot download ROI if Test Setup has TSR AIR and multiple ROIs - Manuscript
 
 
 
 
 
I didn’t write the code in question but it was to allow downloading with multiple ROIs with channels duplicate sensor IDs that come from using voltage input channels on multiple channels, or the embedded channels in a TSR AIR.
 
 
 
 
 
 
                         
                         
       
       
                                           
Dan​​​​
   
 
   
Mcfadden
             
Software Engineering Manager
                     
Diversified Technical Systems, Inc.
                     
e:
       
Dan.Mcfadden@dtsweb.com
       
|
           
o:
       
+1 562 493 0158 x137
       
|
       
www.dtsweb.com
                                         
CONFIDENTIALITY: This email and any attachments may contain information from Diversified Technical Systems, Inc. (DTS) which​ is confidential ​and/or proprietary. If you received this transmission in error, any use, reproduction or dissemination of this transmission​ is strictly prohibited. ​Please delete this message and notify sender immediately.  
 ​EXPORT CONTROL NOTICE: This e-mail may contain technical data whose export, transfer, and/or disclosure may be controlled by​ the US ​International Traffic in Arms Regulation (ITAR) 22 CFR part 120-130 or the Export Administration Regulations (Commerce).
 ​
 ​
                   
 
                         
 
         
   
Kate Martin (DTS)
 
Mar 6, 2025, 16:18 GMT
     
Private note
     
Any idea what this means please ?
  
 2025-03-05 18:20:40.401 aic: Setting channel is configured to False because [SPS00247] CH-01\043270 was compared to [SPS00398] CH-03\053935, [SPS00398] CH-08\052629, [SPS00398] CH-11\048328, [SPS00398] CH-04\053685, [SPS00423] CH-09\054160, [SPS00423] CH-18\054159, [SPS00423] CH-14\053613, [SPS00423] CH-16\052740, [SPS00423] CH-15\053929, [SPS00398] CH-02\048839, [SPS00423] CH-17\053984, [SPS00398] CH-09\051834, [SPS00423] CH-11\053914, [SPS00423] CH-12\053657, [SPS00423] CH-13\053912, [SPS00423] CH-10\053689, [SPS00398] CH-06\050822, [SPS00398] CH-01\051868, [SPS00398] CH-10\051080, [SPS00423] CH-07\053957, [SPS00398] CH-05\054413, [SPS00423] CH-08\054181, [SPS00398] CH-07\051871, [SPS00423] CH-03\053178, [SPS00423] CH-04\049622, [SPS00423] CH-06\049755, [SPS00423] CH-02\053675, [SPS00398] CH-18\039886, [SPS00398] CH-12\038688, [SPS00398] CH-16\039891, [SPS00398] CH-15\049117, [SPS00398] CH-17\039884, [SPS00398] CH-14\031918, [SPS00398] CH-13\054052, [SPS00423]
  
 as it's SPS00247 that has the HV channels in question...
  
 
  
         
 
         
   
Jordon Tray
 
Mar 6, 2025, 11:46 GMT
 
Also the impact was at approx. 18:13.
 
 
 
Regards,
 
 
     
Jordan Tray
 Engineering Technician
     
         
   
   
   
   
   
             
Direct Tel:  
   
+44 24 7635 5000
     
Email:  
   
jordan.tray@horiba-mira.com
         
HORIBA MIRA Ltd.
 Watling Street, Nuneaton
 Warwickshire, CV10 0TU, UK
     
www.horiba-mira.com
 www.vehicletesting.solutions
 www.miratechnologypark.com
     

 If you are visiting one of our sites please use our visitor information guide for Nuneaton  and Quatro Park.
 
Attachment(s)
 image016.jpg
 image017.jpg
         
 
         
   
Jordon Tray
 
Mar 6, 2025, 11:26 GMT
 
Attached as much as possible from yesterday including the setup. I have just been informed this has happened previously on more than 1 test.
 
 
 
Regards,
 
 
     
Jordan Tray
 Engineering Technician
     
         
   
   
   
   
   
             
Direct Tel:  
   
+44 24 7635 5000
     
Email:  
   
jordan.tray@horiba-mira.com
         
HORIBA MIRA Ltd.
 Watling Street, Nuneaton
 Warwickshire, CV10 0TU, UK
     
www.horiba-mira.com
 www.vehicletesting.solutions
 www.miratechnologypark.com
     

 If you are visiting one of our sites please use our visitor information guide for Nuneaton  and Quatro Park.
 
Attachment(s)
 image016.jpg
 image017.jpg
 DASConfigs 2025-03-05 17.45.03.log.bz2
 DASConfigs 2025-03-05 18.14.36.log.bz2
 DASConfigs 2025-03-05 18.17.56.log.bz2
 DASConfigs 2025-03-05 18.20.37.log.bz2
 DASConfigs 2025-03-05 18.26.26.log.bz2
 DASConfigs.log
 DataPRO 2025-03-05 17.37.50.log.bz2
 DASConfigs 2025-03-05 18.16.50.log.bz2
 DataPRO 2025-03-05 18.12.34.log.bz2
 DataPRO 2025-03-05 18.14.30.log.bz2
 DataPRO 2025-03-05 18.18.52.log.bz2
 DataPRO 2025-03-05 18.11.17.log.bz2
 DataPRO 2025-03-05 18.13.32.log.bz2
 DataPRO 2025-03-05 18.26.41.log.bz2
 DataPRO 2025-03-05 18.27.26.log.bz2
 DataPRO 2025-03-05 18.28.10.log.bz2
 DataPRO 2025-03-05 18.21.20.log.bz2
 DataPRO 2025-03-05 18.28.56.log.bz2
 DataPRO 2025-03-05 18.29.40.log.bz2
 DataPRO 2025-03-05 18.30.27.log.bz2
 DataPRO 2025-03-05 18.32.22.log.bz2
 DataPRO 2025-03-05 19.09.32.log.bz2
 DB 2025-03-05 17.38.17.log.bz2
 DataPRO 2025-03-05 18.37.54.log.bz2
 ping.log
 DASConfigs 2025-03-05 17.41.06.log.bz2
 B7455.xml
         
 
         
   
Kate Martin (DTS)
 
Mar 6, 2025, 10:00 GMT
 
Hi Jordon
 
 Can you please attach the test set up and logs ?
 
 Thxs
         
 
         
   
Jordon Tray
 
Mar 6, 2025, 09:46 GMT
 
We have an issue where our HV ROI data from test is cutting off at 45s not 65s as set in the test setup.
         
----------------------------------------------
 Customer Details:
 Name: Jordon Tray ( jordan.tray@horiba-mira.com )
 Organization: Horiba MIRA Ltd
 
----------------------------------------------
 If you no longer wish to see notifications for this ticket, please use the 'Unfollow' option in the Followers List.
 
You are an agent. Add a comment by replying to this email or view ticket in Zendesk Support.
     
Ticket #
   
14994
     
Status
   
On-hold
     
Requester
   
Jordon Tray
     
CCs
   
-
     
Followers
   
Dan McFadden, Jerry Lawrence, Maugan Robert
     
Group
   
Support
     
Assignee
   
Kate Martin
     
Priority
   
Normal
     
Type
   
Problem
     
Channel
   
Web Form
     
 
 
This email is a service from DTS. Delivered by Zendesk.
 
[070XX0-K5L7K]Ticket-Id:14994Account-Subdomain:dtsweb</Comment>
    <StatusString>Opened</StatusString>
    <TimeStamp>2025-03-31T15:32:31Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>Opened</Status>
  </FBEvent>
</ArrayOfFBEvent>