<?xml version="1.0"?>
<ArrayOfFBEvent xmlns:xsd="http://www.w3.org/2001/XMLSchema" xmlns:xsi="http://www.w3.org/2001/XMLSchema-instance">
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment />
    <StatusString>Edited</StatusString>
    <TimeStamp>2023-05-24T20:47:05Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Jaime Lopez</User>
    <Comment>Verified w/ DataPRO version 4.1.197, 4.0.846
 
Using ORCA attached to S6A test device via uart rx (positive) &amp;&amp; uart rx (negative)
 
Created a UART sensor in sensor database and changed data format to NMEA:
 
 
 
 
Using the sensor I created in sensor database I created a test setup and in parameters changed data format to plain text. In system settings I allowed for UART recording. Set recording mode to Circular buffer + UART.
 
 
 
Ran test up to download.
 
Navigating to Data folder -&gt; binary I can see my .txt file created inside</Comment>
    <StatusString>Closed</StatusString>
    <TimeStamp>2023-02-07T19:19:52Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>Closed</Status>
  </FBEvent>
  <FBEvent>
    <User>Jaime Lopez</User>
    <Comment />
    <StatusString>Assigned</StatusString>
    <TimeStamp>2023-01-13T17:41:16Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment />
    <StatusString>Edited</StatusString>
    <TimeStamp>2022-11-30T15:58:40Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment />
    <StatusString>ReleaseNoted</StatusString>
    <TimeStamp>2022-11-29T21:44:43Z</TimeStamp>
    <ReleaseNote>Addresses an issue when downloading with UART where mode was not being respected</ReleaseNote>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>John Dowling</User>
    <Comment>Fix:
 
 
AppendNewUARTDataChunks() was being called with file mode stored in a channel's SensorData rather than the channel itself
 
 
To test:
 
 
Set up a test with a SLICE6AIR with a UART data provider attached (I used a GPS unit). Set the proper settings for the provider in a UART sensor in sensor data and assign that to a UART channel. Use a "+ UART" recording mode. I also assigned an analog channel as well but it wasn't physically hooked up to a source. No clock source assignments are necessary.
 
Under Parameters in Edit Test Setup, click the UART button and change the Data Format for your UART channel to something different that the one assigned in Sensor Database. For instance, I set the format to NMEA in Sensor Database and to Plain Text in Edit Test Setup -&gt; Parameters.
 
Run your test all the way through Download. Check the Binary folder for your test run. The UART data should be stored in the file type that matches your setting in Parameters, not Sensor Database. In my instance, the data was stored to a .txt file (Parameters) rather than .nmea (Sensor Database).
 
 
r47877 for v4.01 dev and v4.00 maint</Comment>
    <StatusString>Resolved</StatusString>
    <TimeStamp>2022-11-22T18:06:42Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>Resolved</Status>
  </FBEvent>
  <FBEvent>
    <User>John Dowling</User>
    <Comment />
    <StatusString>Assigned</StatusString>
    <TimeStamp>2022-11-22T17:11:46Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment />
    <StatusString>Assigned</StatusString>
    <TimeStamp>2022-11-16T21:18:53Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment>Hi Dan
  
 From testing today I think it is something weird going on with the way UART channels are defined in the sensor DB and the default UART config that you get if you first drag and drop a S6A UART channel "TSU_TestSpecific" without having first added your own in the sensor DB.
  
 Two video's uploaded here showing the issues I'm having...
  
 \\dts.local\files\HelpCenter\Private\Zendesk\zd11123
  
 Thxs</Comment>
    <StatusString>Edited</StatusString>
    <TimeStamp>2022-11-16T15:14:39Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment />
    <StatusString>Assigned</StatusString>
    <TimeStamp>2022-11-15T22:07:43Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment>##- Please type your reply above this line -##
 
You are registered as a CC on this support request (dtsweb.zendesk.com/agent/tickets/11123).
 Reply to this email to add a comment to the request.
 
 
 
          
Kate Martin (DTS)
 
Nov 15, 2022, 21:34 GMT
 
Hi Dan
 
 No I defo ran a full test after changing it
 
 Thxs
 
 
         
 
          
Dan McFadden (DTS)
 
Nov 15, 2022, 18:10 GMT
    Private note    
Hi Kate, just to verify, you changed the setting then ran another test and it downloaded as bin, not changed the setting then downloaded again?  
  
 We can try it out here.
 
 
         
 
          
Kate Martin (DTS)
 
Nov 15, 2022, 17:30 GMT
 
Hi Dan
  
 I think I have stumbled on an issue where just changing the serial data file format for the UART is not being respected by DataPro on download.
  
 Originally I had this is binary but changed it here in the test set up to be NMEA ..
  
 
  
  
 and the corresponding das config shows the change ..
  
 
  
 but despite this my data is still .bin 
  
 
 Does the S6A need to be rebooted during run test like it is if the clock sync parameters change ?
  
 
 
 
Attachment(s)
 DataPRO.log
 DataPRO 2022-11-15 14.54.10.log.bz2
 S6A0083.xml
 DASConfigs.log
         
 
 
You are an agent. Add a comment by replying to this email or view ticket in Zendesk Support.
    Ticket # 11123   Status Open   Requester Kate Martin   CCs John Dowling, Dan McFadden, Jerry Lawrence   Group Support   Assignee Kate Martin   Priority Normal   Type Problem   Channel Web Form    
 
 This email is a service from DTS. Delivered by Zendesk.
 
 [6EX2Z4-9D3XL]Ticket-Id:11123Account-Subdomain:dtsweb</Comment>
    <StatusString>Opened</StatusString>
    <TimeStamp>2022-11-15T22:07:42Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>Opened</Status>
  </FBEvent>
</ArrayOfFBEvent>