<?xml version="1.0"?>
<ArrayOfFBEvent xmlns:xsd="http://www.w3.org/2001/XMLSchema" xmlns:xsi="http://www.w3.org/2001/XMLSchema-instance">
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment />
    <StatusString>Edited</StatusString>
    <TimeStamp>2023-05-24T20:44:38Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Justin Richardson</User>
    <Comment>Tested and verified with dp 4 0 717 and 4.1 .43 Diagnostics  and trig check passes and successfully armed the TSR in streaming mode.</Comment>
    <StatusString>Closed</StatusString>
    <TimeStamp>2022-09-16T19:01:37Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>Closed</Status>
  </FBEvent>
  <FBEvent>
    <User>Justin Richardson</User>
    <Comment />
    <StatusString>Assigned</StatusString>
    <TimeStamp>2022-09-16T16:58:14Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment>4.0/4.1@47624</Comment>
    <StatusString>Resolved</StatusString>
    <TimeStamp>2022-09-14T23:22:06Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>Resolved</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment>Pass/Fail, install a new TSR AIR DP, get a brand new TSR AIR (or one initialized to just defaults), create a test setup, run check trigger, it should not automatically be triggered, run diagnostics, it should not normally be failed.</Comment>
    <StatusString>Edited</StatusString>
    <TimeStamp>2022-09-14T23:21:26Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment>should now no longer get clock sync status when enable clock sync is off
 
now should use active mode when Use Circular buffer is set in settings (default), which should fix check trigger auto triggering (as the firmware lies on status)
 
now should not set auto zero deviation % for TSR AIR ever (as the firmware lies on the response).</Comment>
    <StatusString>Edited</StatusString>
    <TimeStamp>2022-09-14T23:11:24Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment />
    <StatusString>Edited</StatusString>
    <TimeStamp>2022-09-14T20:15:38Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment />
    <StatusString>Edited</StatusString>
    <TimeStamp>2022-09-14T20:15:16Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment />
    <StatusString>Assigned</StatusString>
    <TimeStamp>2022-09-14T19:00:27Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment>as reported by CPB, running diagnostics on a TSR with a new install results in a diagnostics that fails with a clock sync error and potentially an error with auto zero deviation.
 
also, check trigger immediately triggered with no level trigger set
 
also, clock sync failure without clock sync enabled</Comment>
    <StatusString>Opened</StatusString>
    <TimeStamp>2022-09-14T19:00:27Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>Opened</Status>
  </FBEvent>
</ArrayOfFBEvent>