<?xml version="1.0"?>
<ArrayOfFBEvent xmlns:xsd="http://www.w3.org/2001/XMLSchema" xmlns:xsi="http://www.w3.org/2001/XMLSchema-instance">
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment />
    <StatusString>Edited</StatusString>
    <TimeStamp>2024-01-02T16:16:57Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Justin Richardson</User>
    <Comment>Tested with fw G2B5    using DataPro  4.2.74 ,   4.0 1025 .. and 4.1.400.49386
1. Enabled Diagnostics mode in dp config file.
2. Created a sensor and checked diagnostic mode ..
3.In realtime measured 2.49v   from exc - to sig + , 
and  2.8v from  exc - to sig -

4. Retested With diagnostics unchecked for sensor  
 measured 50mv  from  exc- to sig +   and  0mv from  exc- to sig - 


RELATED TO CASE 33055 
</Comment>
    <StatusString>Closed</StatusString>
    <TimeStamp>2023-08-16T23:02:16Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>Closed</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment />
    <StatusString>ReleaseNoted</StatusString>
    <TimeStamp>2023-08-15T17:57:04Z</TimeStamp>
    <ReleaseNote>Adds support for Diagnostics Mode for S6A</ReleaseNote>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment />
    <StatusString>Edited</StatusString>
    <TimeStamp>2023-08-15T15:16:28Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment />
    <StatusString>Edited</StatusString>
    <TimeStamp>2023-08-14T16:14:39Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment>4.0/4.1/4.2@49331</Comment>
    <StatusString>Resolved</StatusString>
    <TimeStamp>2023-08-04T18:11:06Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>Resolved</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment>1) turn on support for AllowDiagnosticsMode in DataPRO.exe.config
 
2) set a sensor to have DiagnosticsMode turned on in sensor database
 
 
 
 
3) create a test setup with that sensor
 
4) it should collect flatline data regardless of whether a sensor is attached or not, you should see diagnostics mode in the UI
 
 
in viewing data you'll still see noise (around 18ADC maybe) but it should be stable.  You can also check for the command being set, it will set voltage insertion when configuring, then turn it off again (by sending an empty list of channels) after the event is complete.
 
2023-08-04 10:46:10.720 -&gt;  Diagnostics - SetChannelDiagnosticVoltageInsertion_SLICE2 [00000164], S6A0275, local: 192.168.3.83:52536, Remote: 192.168.4.75:8301 StackChannels: (0, 2)</Comment>
    <StatusString>Edited</StatusString>
    <TimeStamp>2023-08-04T17:56:36Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment />
    <StatusString>Assigned</StatusString>
    <TimeStamp>2023-08-04T17:15:24Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment>per LP S6A has diagnostics mode support and it should be enabled for this das (but only S6A, not S6, etc)</Comment>
    <StatusString>Opened</StatusString>
    <TimeStamp>2023-08-04T17:15:24Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>Opened</Status>
  </FBEvent>
</ArrayOfFBEvent>