<?xml version="1.0"?>
<ArrayOfFBEvent xmlns:xsd="http://www.w3.org/2001/XMLSchema" xmlns:xsi="http://www.w3.org/2001/XMLSchema-instance">
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment />
    <StatusString>Edited</StatusString>
    <TimeStamp>2024-08-24T00:13:33Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Justin Richardson</User>
    <Comment>Verified with Datapro Versions 4.0.1152.50700, 4.2.418.50819 and 4.3.165.50821


Tested using a S6DB and system of s6  
- Armed system of s6das with s6db and shorted status  on various ports to verify  datapro shows the correct extended fault for that port.
-Checked arm attr 127 extended faults  flags  after fault was set and verified the Datapro showed the correct fault for the  bit that was  set







</Comment>
    <StatusString>Closed</StatusString>
    <TimeStamp>2024-05-23T21:23:00Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>Closed</Status>
  </FBEvent>
  <FBEvent>
    <User>Justin Richardson</User>
    <Comment>I think you meant checked in at  4.0/4.2/4.3    , Jerry Confirmed this...
</Comment>
    <StatusString>Edited</StatusString>
    <TimeStamp>2024-05-23T20:39:45Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Justin Richardson</User>
    <Comment />
    <StatusString>Assigned</StatusString>
    <TimeStamp>2024-05-23T20:15:23Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment />
    <StatusString>Edited</StatusString>
    <TimeStamp>2024-05-07T17:39:35Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment />
    <StatusString>Edited</StatusString>
    <TimeStamp>2024-05-07T17:39:30Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment />
    <StatusString>ReleaseNoted</StatusString>
    <TimeStamp>2024-05-06T23:46:10Z</TimeStamp>
    <ReleaseNote>Corrects an issue where S6DB extended faults were not being interpreted correctly</ReleaseNote>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment />
    <StatusString>ReleaseNoted</StatusString>
    <TimeStamp>2024-05-06T23:45:50Z</TimeStamp>
    <ReleaseNote>S6DB extended faults were not being interpreted correctly</ReleaseNote>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment>4.0/4.1/4.2@50663</Comment>
    <StatusString>Resolved</StatusString>
    <TimeStamp>2024-04-29T23:08:56Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>Resolved</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment />
    <StatusString>Assigned</StatusString>
    <TimeStamp>2024-04-29T22:52:31Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Justin Richardson</User>
    <Comment />
    <StatusString>Edited</StatusString>
    <TimeStamp>2024-04-25T17:20:01Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Justin Richardson</User>
    <Comment />
    <StatusString>Edited</StatusString>
    <TimeStamp>2024-04-25T17:19:33Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Justin Richardson</User>
    <Comment />
    <StatusString>Edited</StatusString>
    <TimeStamp>2024-04-25T17:18:59Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment>also apply this to 4.0.</Comment>
    <StatusString>Edited</StatusString>
    <TimeStamp>2024-04-25T17:18:37Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Dan McFadden</User>
    <Comment />
    <StatusString>Edited</StatusString>
    <TimeStamp>2024-04-25T17:07:52Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Justin Richardson</User>
    <Comment />
    <StatusString>Edited</StatusString>
    <TimeStamp>2024-04-25T16:22:11Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Justin Richardson</User>
    <Comment />
    <StatusString>Edited</StatusString>
    <TimeStamp>2024-04-25T16:21:39Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Justin Richardson</User>
    <Comment />
    <StatusString>Edited</StatusString>
    <TimeStamp>2024-04-25T16:21:30Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Justin Richardson</User>
    <Comment />
    <StatusString>Edited</StatusString>
    <TimeStamp>2024-04-25T16:21:08Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Justin Richardson</User>
    <Comment />
    <StatusString>Edited</StatusString>
    <TimeStamp>2024-04-25T16:20:56Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Justin Richardson</User>
    <Comment />
    <StatusString>Edited</StatusString>
    <TimeStamp>2024-04-25T16:09:02Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Justin Richardson</User>
    <Comment />
    <StatusString>Edited</StatusString>
    <TimeStamp>2024-04-24T21:21:27Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Justin Richardson</User>
    <Comment />
    <StatusString>Edited</StatusString>
    <TimeStamp>2024-04-24T21:20:53Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Justin Richardson</User>
    <Comment />
    <StatusString>Assigned</StatusString>
    <TimeStamp>2024-04-24T20:53:38Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>UNKNOWN</Status>
  </FBEvent>
  <FBEvent>
    <User>Justin Richardson</User>
    <Comment>Extended faults from S6DB  need to be updated  to match fwtu ..
Datapro version 4.2.353  and s6db  fw B0N5

Noticed this issue when testing S6db  fw ,  that extended faults are offset by 1 in datapro .

-  If I arm s6db with s6 das attached to port 1 and then   short the status to gnd  the das gives an extended fault code = 1 for arm attribute 127  which should show in datapro as a extended fault for dropped status on port 1 but it does not show any extended faults.

- If I  short the status on port 2 the das gives fault code =2  for arm attribute 127  and datpro shows extended fault  status dropped for port 1   not port 2 as expected.  Extended faults are offset by 1. 

- Opened a case for fw and this  is the correct behavior for fw  and fwtu was updated to accomodate it , per Edward Datapro needs to also be updated   to display  extended  fault flags.


 
see case 11020 

pasted below is from the datapro  log showing  an extended fault for port 2 ( I shorted status line on port 2)   but  in the log is  showing port 1 ...







2024-04-24 13:37:19.770 &lt;-  Arm - QueryArmAndTriggerStatus [00048260]\[00048260] - (1.0056ms), S6DB0032 Event number: 0, Arm Mode: 1 Armed: True, Started: True, Triggered: False, Faulted: True, Diagnostics: False, FlashClear: False, Arming: False, ArmState: 1 Samplerate: 10000, Current Sample: 10000, Total Samples: 211000, Fault Sample Number: 0 Input Voltage: 16.474V, Backup Voltage: 0.000V, Backup Percentage Remaining: 0% System Temp: 36.98263C, Actual Current Sample: 10000, Max Sample Available: 211000 Tilt Ch1: 0, Tilt Ch2: 0, Tilt Ch3: 0 Squib Ch1: 0, Squib Ch2: 0, Squib Ch3: 0, Squib Ch4: 0 Squib P17V: 0, Squib PV5: 0, Squib P3p3V: 0, Squib N5V: 0 PTP Enabled: False, PTP Clk Enable: False, PTP Clk Sync Status: False, ADC Clk Sync Status: True Fault Flags: IncomingStatusLineDropped, InputVoltageLow Extended Fault Flags: EXT_FAULT_TYPE_STATUSLINE_PORT_1_DROPPED

</Comment>
    <StatusString>Opened</StatusString>
    <TimeStamp>2024-04-24T20:53:37Z</TimeStamp>
    <ZendeskNumber>0</ZendeskNumber>
    <Status>Opened</Status>
  </FBEvent>
</ArrayOfFBEvent>